title: Volume-displacement body plethysmograph using a large flowmeter without pressure compensation.
abstract: A modification of the conventional volume-displacement whole-body plethysmograph was done using a low-resistance flow element and integrating the plethysmographic pressure signal. A flow element with a resistance of 0.036 cmH2O/(l/s), linear to 75 l/s was fitted to a 440-liter plethysmograph, resulting in an instrument with a volume-amplitude-frequency response flat to 15 Hz and in phase with mouth flow and esophageal catheter-balloon to at least 15 Hz without pressure compensation.
mesh_terms: D006801:Humans; D008168:Lung / Q000502:physiology; D010993:Plethysmography, Whole Body / Q000295:instrumentation*; D011312:Pressure; D012212:Rheology