Evaluation of microwave hyperthermia applicators.
Hyperthermia has been used in conjunction with radiation and chemotherapy for cancer treatment. When using electromagnetic heating, applicators are critical components in contact with or in proximity to patients and can be the determining factor for effective and safe treatment. Tissue absorption of electromagnetic energy is determined by many factors. Three cases are shown to illustrate the complexity of microwave heating: 1) The BSD MA-151 applicator has good center heating on a muscle-only phantom as shown in the operation manual. When fat slabs of 0.25, 0.5, 1, and 2 cm thick were added, two hot spots near the periphery of the applicator were evident on all fat surfaces, exposed at 631 MHz. At 915 MHz, the heating was elongated on the surface of the models with 0.25- and 2-cm fat, and two hot spots were observed on the 0.5- and 1-cm fat surfaces. 2) Heating patterns of the Clini-Therm applicators on a muscle-only phantom, as indicated in the operations guide, are elliptical with their major axes perpendicular to the electric field. However, when a bolus is used, the elliptical pattern is parallel to the E field. 3) Heating patterns in cylindrical structures were studied with inhomogeneous models of limbs. Arm and thigh models consisting of fat, bone, and muscle material were heated with Clini-Therm L, M, and MS applicators at 915 MHz. In addition to the geometric effect, the results indicated that placing the applicators with E field parallel to the long axis of cylindrical structures can minimize required power, produce less heating of fats and reduce stray radiation. In conclusion, to apply penetrating microwave or other RF fields for tissue heating, one must simulate the clinical exposure conditions as closely as possible to obtain useful heating patterns.
Chou
C K
CK
Department of Radiation Research, City of Hope National Medical Center, Duarte, California 91010.
D005069:Evaluation Studies as Topic; D006801:Humans; D006979:Hyperthermia, Induced / Q000295:instrumentation*; D008872:Microwaves; D008961:Models, Structural; D013817:Thermography