title: [An unusual cause of anesthetic ventilator malfunction].
abstract: We report a case of malfunction of an anesthetic ventilator by an unusual cause. A 48-year-old male with gastric cancer was scheduled for gastrectomy. Anesthesia was maintained with enflurane, N2O, O2 and epidural blockade using a semiclosed circuit system. The patient was ventilated using AV1 anesthetic machine (Dräger Co.). Forty minutes after induction of anesthesia, chest movement of the patient suddenly stopped. There was no gas flow to the patient during inspiratory phase. Air leak was not found in anesthetic respiratory circuit and at the bellows of the ventilator. The supply of oxygen and air to the anesthetic machine was sufficient. Since we could not find any cause of the ventilator failure, anesthesia was maintained with manual ventilation by using another anesthetic machine until completion of the surgery. After the surgery, we recognized that the controller unit of expiratory valve of the ventilator was obstructed by a Tamper Proof Film, which seals the outlet of a commercial bag of lactated Ringer's solution (Solulact, Terumo Co.). It seems that the film dropped accidentally between the main part and the ventilator system of anesthetic machine when the bellows was exchanged before the surgery, and moved on to the controller unit of the expiratory valve of the ventilatory system during surgery. In conclusion, it is necessary for anesthetists to understand the inner structure and system of the anesthetic machine and to check the anesthetic machine to avoid the troubles and accidents related to anesthetic machine.
authors: Division of Anesthesia, Asahikawa City General Hospital.
authors: M
authors: M
authors: Kawamata
authors: T
authors: T
authors: Mayumi
authors: M
authors: M
authors: Miyabe
authors: A
authors: A
authors: Namiki
mesh_terms: D000764:Anesthesia, Closed-Circuit / Q000295:instrumentation*; D004868:Equipment Failure; D006801:Humans; D008297:Male; D008875:Middle Aged; D012122:Ventilators, Mechanical