Two-dimensional color-mapping of turbulent shear stress distribution downstream of two aortic bioprosthetic valves in vitro.
Since artificial heart valve related complications such as thrombus formation, hemolysis and calcification are considered related to flow disturbances caused by the inserted valve, a thorough hemodynamic characterization of heart valve prostheses is essential. In a pulsatile flow model, fluid velocities were measured one diameter downstream of a Hancock Porcine (HAPO) and a Ionescu-Shiley Pericardial Standard (ISPS) aortic valve. Hot-film anemometry (HFA) was used for velocity measurements at 41 points in the cross-sectional area of the ascending aorta. Three-dimensional visualization of the velocity profiles, at 100 different instants during one mean pump cycle, was performed. Turbulence analysis was performed as a function of time by calculating the axial turbulence energy within 50 ms overlapping time windows during the systole. The turbulent shear stresses were estimated by using the correlation equation between Reynolds normal stress and turbulent (Reynolds) shear stress. The turbulent shear stress distribution was visualized by two-dimensional color-mapping at different instants during one mean pump cycle. Based on the velocity profiles and the turbulent shear stress distribution, a relative blood damage index (RBDI) was calculated. It has the feature of combining the magnitude and exposure time of the estimated shear stresses in one index, covering the entire cross-sectional area. The HAPO valve showed a skewed jet-type velocity profile with the highest velocities towards the left posterior aortic wall. The ISPS valve revealed a more parabolic-shaped velocity profile during systole. The turbulent shear stresses were highest in areas of high or rapidly changing velocity gradients. For the HAPO valve the maximum estimated turbulent shear stress was 194 N m-2 and for the ISPS valve 154 Nm-2. The RBDI was the same for the two valves. The turbulent shear stresses had magnitudes and exposure times that might cause endothelial damage and sublethal or lethal damage to blood corpuscules. The RBDI makes comparison between different heart valves easier and may prove important when making correlation with clinical observations.
Nygaard
H
H
Engineering College, Aarhus, Denmark.
Giersiepen
M
M
Hasenkam
J M
JM
Reul
H
H
Paulsen
P K
PK
Rovsing
P E
PE
Westphal
D
D
D001021:Aortic Valve; D001705:Bioprosthesis; D001783:Blood Flow Velocity / Q000502:physiology; D001792:Blood Platelets / Q000502:physiology; D001794:Blood Pressure / Q000502:physiology; D002302:Cardiac Output / Q000502:physiology; D003116:Color; D006350:Heart Valve Prosthesis; D006439:Hemodynamics / Q000502:physiology; D006801:Humans; D008955:Models, Cardiovascular; D015539:Platelet Activation / Q000502:physiology; D011474:Prosthesis Design; D011673:Pulsatile Flow; D012039:Regional Blood Flow / Q000502:physiology; D012212:Rheology; D013314:Stress, Mechanical