Impulse noise generator--design and operation.
In the seventies PFANDER (Pfander, 1975) proposed a screening test with an impulse noise simulator to check the particular responsivity of soldiers on vulnerability of the inner ear concerning the impulse noise-induced hearing loss. According to a system developed at the University of Oldenburg (Germany) (Klug & Radek, 1987), we have constructed an impulse noise generator designed for our specific requirements that will be presented. The simulator consists of an electrical ignited impulse noise spark gap which is supplied by a 3.5 kV high voltage source. At a distance of 1.10 m from the center of the impulse noise spark gap a peak pressure level of 155 dB with a C-Duration (Pfander, 1975) of .2 msec and with the main energy in the frequency range from 1 kHz to 2 kHz was good reproducible. It would be preferable to shift the impulse noise spectrum to lower frequencies but experimental effort has failed so far.
Brinkmann
H
H
Wehrtechnische Dienststelle für Waffen und Munition, Meppen, Germany.
D000161:Acoustic Stimulation / Q000295:instrumentation; D001299:Audiometry / Q000295:instrumentation*; D004867:Equipment Design; D005260:Female; D006317:Hearing Loss, Noise-Induced / Q000175:diagnosis* / Q000209:etiology; D006801:Humans; D008297:Male; D008889:Military Personnel; D009622:Noise / Q000009:adverse effects