Disinfection of endoscopic equipment.
Patients undergoing endoscopy are at risk of infection from the use of contaminated equipment. Dangers arise from the transmission of organisms from one patient to another and from the introduction of opportunist organisms which colonize endoscopic equipment on storage and can lead to sepsis and death in those who are immunocompromised and at ERCP. Staff are in danger from needle-stick injury and sensitivity to aldehyde disinfectants. These risks can be eliminated by careful attention to disinfection techniques. The most important part of endoscope disinfection is thorough mechanical cleaning first, followed by 5-10 min total immersion of the instrument and all channels in 2% glutaraldehyde (or the equivalent). At the end of the endoscopy list, following the disinfection protocol, all equipment should be dried internally and externally prior to storage. Staff must be fully aware of the risks of infection in endoscopy, be protected from hepatitis B by vaccination, and be fully trained in disinfection techniques. Glutaraldehyde should be used only in closed systems or in well-ventilated areas with the operator protected from direct contact from splashing and fumes. Institutions should designate an individual to be responsible for preparing, monitoring and overseeing disinfection procedures within the endoscopy room and for ensuring that regular microbiological testing of equipment (including automatic disinfecting machines) is undertaken.
Axon
A T
AT
D004203:Disinfection / Q000295:instrumentation / Q000379:methods*; D020776:Endoscopes, Gastrointestinal; D004866:Equipment Contamination / Q000517:prevention & control*; D005976:Glutaral / Q000008:administration & dosage; D006801:Humans
D005976:Glutaral