[Formation of a knot in a J spiral metallic guide: a complication of the Seldinger method].
Central venous catheter insertion by Seldinger's technique uses a wire guide which could be cause of complications. This case reports a blocked wire guide of which J tip was responsible for knotting. Utilization of vessel dilator of central venous catheterization set (Arrow) permitted to liberate and withdraw the wire guide without surgical operation. A larger use of this wire guide could increase the frequency of such complications.
Carpentier
J P
JP
Service d'Anesthésie-Réanimation, CHA René-Le-Bas, Cherbourg-Naval.
Braz da Silva
J
J
Choukroun
G
G
D000368:Aged; D002405:Catheterization, Central Venous / Q000009:adverse effects* / Q000295:instrumentation / Q000379:methods; D006801:Humans; D008297:Male
Formation d'un noeud sur un guide métallique spiralé en J: une complication de la méthode de Seldinger.