Prevention of oral bacterial flora transmission by using mouth-to-mask ventilation during CPR.
The Emergency Cardiac Care Committee of the American Heart Association has recently recommended utilizing protective barrier precautions during CPR (1,2). We assessed 17 mask and faceshield resuscitation devices for adequacy of barrier protection. Eight of the devices were faceshields (CPR Microshield, Hygenic, MedCare Mask, Resusci, Samaritan, Sealeasy, Portex); 8 were mask devices (Laerdal, Dyna Med, MTM Emergency Lung Ventilator, MTM Emergency Resuscitator, Res-Q-Flo, Rightway Mouth-to-Mask Resuscitation, Trufit), and one of the devices did not meet the criteria for either faceshield or mask (Lifesaver). All masks were disinfected, applied to the investigator's face as directed by the manufacturers' instructions, and then cultured for oral aerobic bacterial flora on the rescuer side. No mask devices cultured positive for oral aerobic bacterial flora, while 6 of 8 faceshield devices cultured positive for oral aerobic bacterial flora (P less than 0.007). The CPR Microshield and the Portex faceshield were the only devices that did not develop a positive culture. We conclude that all ventilation devices with a one-way valve, except the Sealeasy device, provide adequate barrier type protection from oral aerobic bacterial flora when simulating mouth-to-barrier type protection when performing mouth-to-mouth ventilation.
Cydulka
R K
RK
Department of Emergency Medicine, MetroHealth Medical Center, Cleveland, OH 44109.
Connor
P J
PJ
Myers
T F
TF
Pavza
G
G
Parker
M
M
D003428:Cross Infection / Q000517:prevention & control; D004867:Equipment Design; D006323:Heart Arrest / Q000628:therapy*; D006801:Humans; D007239:Infections / Q000635:transmission*; D008397:Masks; D009055:Mouth / Q000382:microbiology*; D012121:Respiration, Artificial / Q000295:instrumentation* / Q000379:methods; D012151:Resuscitation