A high-yield sampler for toxicological characterization of complex mixtures in combustion effluents.
Combustion sampling for toxicological assessment often requires that large (greater than 100 mg) lots of complex organic mixtures of wide volatility range be rapidly recovered from high temperature gases without contamination. A new sampler, meeting these criteria for studies of public health interest, has been developed and demonstrated. The device provides high sampling rates and intimate contacting of the samples stream with large volumes of a well-cooled, liquid solvent, dichloromethane (DCM). This promotes rapid organics dissolution from carrier gas and particulates and prompt dilution and quenching of the resulting solution, resulting in high organics collection efficiencies with minimal DCM losses. Solvent separation then remits large quantities of concentrated organics for chemical analysis and toxicological testing. One- to seven-hour interrogations of in-flame, post-flame, and flue gas regions gave 50- to 250-mg yields of complex organic mixtures. In side-by-side sampling of combustion exhaust, the DCM sampler provided higher yields of DCM solubles (identified with complex organic mixtures) and of S. typhimuirim mutagens (active without exogenous metabolizing agents) than did a filter/polymeric sorbent bed sampling train. The new sampler also collects polar and high volatile hydrocarbons such as benzaheyde, pentadiyne, m- and p-diethynyl-benzene, and 1-hexen-3,5-diyne. Nitration of naphthalene and pyrene in DCM solution (1 mg/mL each) was less than 1 part in 10(7) after a 345-min exposure to a bubbling flow of moist N2/air mixture (1:1 v/v) containing 107 ppm NO and 1.5 ppm NO2, indicating that for these condition a DCM sampler should resist artifactual nitration of aromatics. However, because of the very high bacterial mutagenicity of some nitroaromatics and the wide range of sampling conditions of environmental interest, nitration and all artifacts must still be scrutinized when using the DCM sampler. The DCM sampler is expected to contribute to public health impact assessments by facilitating detailed determinations of the identities, compositions, concentrations, sources, formation mechanisms, and biological activity of environmental toxicants in gaseous atmospheres.
Kruzel
E L
EL
Energy Laboratory, Massachusetts Institute of Technology, Cambridge 02139.
Lafleur
A L
AL
Braun
A G
AG
Longwell
J P
JP
Thilly
W G
WG
Peters
W A
WA
D001274:Atmospheric Pressure; D004784:Environmental Monitoring / Q000295:instrumentation*; D004867:Equipment Design; D005374:Filtration / Q000295:instrumentation; D006256:Hazardous Waste; D009153:Mutagens / Q000032:analysis; D009566:Nitrates / Q000032:analysis; D011083:Polycyclic Compounds / Q000032:analysis; D011108:Polymers; D012015:Reference Standards; D012037:Refuse Disposal / Q000379:methods*; D012997:Solvents; D014835:Volatilization
D006256:Hazardous Waste; D009153:Mutagens; D009566:Nitrates; D011083:Polycyclic Compounds; D011108:Polymers; D012997:Solvents