Electric impedance cuff for the indirect measurement of blood pressure and volume elastic modulus in human limb and finger arteries.
A new plethysmograph, the electric impedance cuff, was designed for the indirect measurement of blood pressure, volume elastic modulus Ev and compliance Ca in human limb arteries. This comprises a compression chamber filled with electrolyte solution and a tetrapolar electric impedance plethysmograph whose electrodes are placed inside the chamber; the former for controlling transmural arterial pressure Pt, and the latter for detecting total limb volume Vo, mean arterial volume Va and its variation delta Va. Systolic and mean arterial pressure in the upper arms, forearms and fingers were measured by detecting pulsatile impedance variation during the gradual (3-5 mm Hg per heart beat) increase (or decrease) in chamber pressure by the volume oscillometric technique. Diastolic and pulse pressure delta P were calculated from these pressure values. Compliance Ca = delta V/delta P and volume elastic modulus Ev = delta P/(delta Va/Va) were recorded at various Pt levels, controlled by the compression pressure. Although this is a kind of impedance plethysmograph, the volume change in a limb segment can be detected by this method without passing electric current through the limb.
Shimazu
H
H
Kawarada
A
A
Ito
H
H
Yamakoshi
K
K
D000328:Adult; D001158:Arteries / Q000502:physiology; D001795:Blood Pressure Determination / Q000295:instrumentation*; D004548:Elasticity; D004867:Equipment Design; D005385:Fingers / Q000098:blood supply*; D005542:Forearm / Q000098:blood supply*; D006801:Humans; D008297:Male; D010992:Plethysmography, Impedance / Q000295:instrumentation*; D011673:Pulsatile Flow; D012016:Reference Values