Nondestructive, three-dimensional internal fit mapping of fixed prostheses.
Crown fit may be three-dimensionally mapped by using a nondestructive optical technique. Reflected light transmission through thicknesses of a colored impression material was shown to follow the Beer-Lambert relationship over a range of approximately 10 to 300 microns, a useful range of interest for study of cement layer thickness. Control thicknesses of impression material were first formed between the measuring platens of a micrometer, and light transmission values (relative reflections) were measured through these control thicknesses of impression material held against air-abraded, noncast gold alloy. Relative reflection values were similarly measured from impression material formed inside cast gold crowns seated on their respective dies and these values were plotted against direct thickness measurements by using the Beer-Lambert relationship. It was established that the micrometer-produced control specimens yielded a valid standard curve for use with the cast gold crowns. Precementation space measurements determined photometrically were found to correlate significantly (r = 0.72, p less than 0.001) with zinc phosphate cement thickness measured on subsequently cemented and cross-sectioned crowns. A three-dimensional representation of the precementation space inside one crown was developed after mapping of 45 points in less than 5 minutes.
Kelly
J R
JR
Harvard School of Dental Medicine, Boston, Mass.
Davis
S H
SH
Campbell
S D
SD
D002484:Cementation; D003442:Crowns; D003760:Dental Impression Materials; D003779:Denture Design; D003781:Denture Retention; D010783:Photometry; D013499:Surface Properties
D003760:Dental Impression Materials