European multicentre evaluation of the ABBOTT Spectrum clinical chemistry analyzer.
The analytical performance of the selective multitest ABBOTT Spectrum analyser was studied according to the ECCLS guidelines and partly the CERMAB protocol in a multicentre evaluation involving laboratories from six European countries. Fifteen analytes, including the electrolytes sodium, potassium and chloride, were measured each in at least 3 laboratories, all at 37 degrees C, except the electrolytes, which are measured at room temperature. The trial lasted approximately three months and involved the collection of over 60,000 data points. It yielded the following results: 1. The precision was at least as good as the precision obtained with the comparison instruments. The majority of the coefficients of variation were between 1 and 4%. 2. The recovery for method assigned control sera values was, with few exceptions, within 10%. 3. Good agreement with respect to the method assigned values of control materials and method comparison with patient specimens to different instruments (e.g. SMAC, Hitachi 737, RA 1000) was found. 4. No drift was observed. 5. Reagent-related carry-over was not found. Specimen-related carry-over was detected in some cases, the deviation being of little or no clinical significance. 6. The manufacturer's claims regarding method linearity were as stated or exceeded. 7. The open system capability was tested and rated as very convenient. 8. The practicability of the instrument was very good.
Blijenberg
B G
BG
Akademisch Ziekenhuis, Rotterdam, The Netherlands.
Braconnier
F
F
Vallez
J M
JM
Burlina
A
A
Plebani
M
M
Celadin
M
M
Haeckel
R
R
Römer
M
M
Hänseler
E
E
De Schrijver
G
G
D001774:Blood Chemical Analysis / Q000295:instrumentation*; D005060:Europe; D005069:Evaluation Studies as Topic; D006801:Humans; D015337:Multicenter Studies as Topic