A comparative study of BP and USP rotating basket dissolution apparatus.
The difference between BP and USP rotating basket dissolution apparatus was investigated by applying both methods to a range of commercial formulations and the NCDA Performance Standard II. No significant differences were obtained between the two sets of apparatus for the commercial tablets. However appreciable difference was observed with the NCDA calibrator.
Smith
W J
WJ
Heaume
P E
PE
Hailey
D M
DM
Lea
A R
AR
D010602:Pharmacopoeias as Topic; D012995:Solubility; D013607:Tablets; D006113:United Kingdom; D014481:United States
D013607:Tablets