On-line computer estimation of carbon dioxide response curves.
Anesthesiologists are concerned with the effect of various anesthetics on a patient's central nervous ventilatory control. The most widely accepted method of determining the effect of a drug is to compare carbon dioxide response curves (delta VE/delta PETCO2, where VE = minute ventilation [in L/min] and PETCO2 = end-tidal carbon dioxide [in mm Hg]) measured before and after administration of the drug. Additional information concerning neuromechanical control can be obtained by also including a measure of the airway occlusion pressure (generally measured 100 ms after occlusion, i.e., P100). To facilitate these measurements we have developed a portable, computer-controlled data acquisition system. It includes an Apple II+ computer and measures VE, PETCO2, and P100. Each subject rebreathes exhaled carbon dioxide through a two-way breathing valve attached to a 9-liter reservoir, which is initially filled with 5% carbon dioxide and balance oxygen. Exhaled carbon dioxide concentrations are measured with an infrared medical gas analyzer on samples taken through a catheter connected at the mouthpiece. The exhaled flow is measured with a pneumotachograph in conjunction with a differential pressure transducer, and P100 is determined with a Validyne MP45 pressure transducer.
D000403:Airway Resistance; D000758:Anesthesia; D002245:Carbon Dioxide / Q000032:analysis / Q000502:physiology*; D002490:Central Nervous System / Q000502:physiology; D006801:Humans; D008297:Male; D008838:Microcomputers; D008991:Monitoring, Physiologic; D009862:Online Systems; D012123:Pulmonary Ventilation; D012119:Respiration; D013147:Spirometry
D002245:Carbon Dioxide