Water pressure and flow regulation for water-cooled lasers.
We experienced laser water valve failure resulting from poor water quality, frequent laser shutdowns from low water flow rates, and unnecessary service calls shortly after installing a new laser. The water valve failure resulted from deposits and corrosion. A dirt/rust water filter was installed, and no further water valve failure has occurred. A flow meter was added to the water system to adjust flow rates. It clearly shows when laser shutdowns are caused by low flow rates and indicates the need for water filter changes. Water pressure was monitored and is most affected by use of the laser. A convenient electric water control, activated by the laser key switch, has proved to be reliable. The water control is kept open by a timer ten minutes after the laser is shut off. We determined that our laser shutdowns were related to transient drops in water flow rates and possibly to draw off of water in other parts of the hospital.
Chambers
J K
JK
Storm Eye Institute, Medical University of South Carolina, Charleston 29425.
Talansky
M L
ML
D003080:Cold Temperature; D004559:Electric Wiring; D004867:Equipment Design; D004868:Equipment Failure; D005374:Filtration; D006874:Hydrostatic Pressure; D007834:Lasers; D011312:Pressure; D012212:Rheology; D012498:Sanitary Engineering; D014873:Water Pollutants
D014873:Water Pollutants