A self-sealing dialysis prosthesis. Coaxial double PTFE-silicone graft.
The performance of a new dialysis prosthesis designed to self-seal after puncture was tested ex vivo and in vivo. It consists of two coaxial polytetrafluoroethylene tubes (PTFE), the space between them filled with silicone rubber sealant (PTFE-sil). Ex vivo: Three PTFE-sil, three double PTFE (without silicone), and three single PTFE grafts were placed sequentially between scribner shunts in the hind limb of four dogs. Bleeding on puncturing with an 18-gauge needle was measured for 30 seconds. PTFE-sil bled less than the controls (g): PTFE-sil; 16 +/- 18; double PTFE: 32 +/- 10; single PTFE: 52 +/- 19 (p less than 0.001). In vivo: Six PTFE-sil and five single PTFE grafts were interposed between the carotid artery and jugular vein of dogs and were punctured with a 16-gauge needle on days 1, 3, and 7. Bleeding was measured through an incision over the puncture site at 5 minutes. In 11 punctures of PTFE-sil, there was no bleeding; three bled less than 20 g. In 13 control punctures, none bled less than 70 g. Patency: Grafts were studied for patency in arteriovenous (AV) fistulas in 34 dogs. Each dog received a PTFE-sil graft in one groin and a single PTFE control graft in the other. At 4 months, patency rates were: PTFE-sil, 84%; single PTFE, 87% (NS). Four months after implantation, hemostasis after puncture in PTFE-sil grafts required 70 +/- 49 seconds versus 207 +/- 48 seconds for PTFE grafts (p less than 0.005). In conclusion, PTFE-sil grafts are self-sealing, can be used immediately after implantation, and need minimal compression after needle removal.
D000818:Animals; D001807:Blood Vessel Prosthesis; D003956:Dialysis / Q000295:instrumentation*; D004285:Dogs; D005069:Evaluation Studies as Topic; D006083:Graft Occlusion, Vascular / Q000473:pathology; D007688:Kidneys, Artificial; D010477:Perfusion; D011138:Polytetrafluoroethylene; D019736:Prostheses and Implants; D011677:Punctures
D011138:Polytetrafluoroethylene