Optimization of ultrasonic transducers.
The so-called KLM-model for ultrasonic transducers is employed to optimize transducer design. Some new performance characteristics are defined which change monotonically with design parameters. These characteristics are based on the area of the envelope of the echo waveform produced by the transducer and of the corresponding amplitude spectrum. The efficiency of the transducer is defined by the round trip energy factor. The performance characteristics are used in a composite performance measure, which is then employed as a criterion in the optimization procedure. Two transducers are investigated: for medical imaging purposes and for spectral analysis of clinical echograms. The influence of electrical matching, backing impedance, matching layer impedance, bond line thickness and series induction on the optimized transducers is investigated.
D004867:Equipment Design; D006801:Humans; D014159:Transducers; D014463:Ultrasonography / Q000295:instrumentation*