Left ventricular outflow tract obstruction with mitral valve replacement in small ventricular cavities.
The inference that mitral valve replacement (MVR) may produce left ventricular outflow tract (LVOT) obstruction has been made, but no comparative hemodynamic studies with various types of prostheses have been done. The purpose of the present study was to compare the gradients created across the LVOT with MVR in young sheep with small left ventricular cavities. Mitral valve replacement was accomplished using cardiopulmonary bypass and hypothermic cardioplegic arrest. Five animals were used for each of the following valves studied: 25-mm Ionescu-Shiley bovine pericardial valve, 25-mm Hancock porcine aortic valve, 2M-6120 28-mm Starr-Edwards ball-valve prosthesis, 25-mm Björk-Shiley 60-degree flat tilting-disc prosthesis, and 25-mm St. Jude Medical hemidisc valve. Gradients across the LVOT were measured after MVR and then during infusion of isoproterenol hydrochloride (0.05 micrograms/kg/min). Following MVR, only the Starr-Edwards valve produced an LVOT gradient (32 +/- 23 mm Hg). Substantial gradients after MVR were seen, however, with isoproterenol administration with the Ionescu-Shiley (47 +/- 4 mm Hg), Hancock (13 +/- 8 mm Hg), and Starr-Edwards (65 +/- 30 mm Hg) valves but not with the low-profile valves (Björk-Shiley and St. Jude Medical). The results of the present study demonstrate that MVR can produce LVOT obstruction. The greatest degree of obstruction was with the high-profile mechanical and bioprosthetic valves.
D000818:Animals; D001705:Bioprosthesis / Q000009:adverse effects; D006350:Heart Valve Prosthesis / Q000009:adverse effects*; D006352:Heart Ventricles / Q000033:anatomy & histology* / Q000503:physiopathology; D006439:Hemodynamics; D008943:Mitral Valve / Q000503:physiopathology* / Q000601:surgery; D012756:Sheep; D013318:Stroke Volume