Hemodynamic evaluation of the integral monostrut Björk-Shiley prosthesis in the aortic position.
Between May, 1983, and November, 1984, the new integral monostrut Björk-Shiley prosthesis was used for aortic valve replacement in 62 patients. The prosthesis is machined from a solid piece of cobalt alloy and has no welded joints. The traditional U-shaped outlet strut has been replaced by a projecting metal finger that holds the disc in place. The disc opens to 70 degrees and is convexoconcave. Successful transseptal heart catheterization was performed in 23 patients an average of 6 months following operation to evaluate the hemodynamic performance of the prosthesis. The mean peak-to-peak gradient was 7.73 +/- 7.49 mm Hg (+/- standard deviation). In five valves it was 0, and in only three was it higher than 15 mm Hg. Significant peak gradients were directly related to the valve index (valve area/body surface area). Mean systolic gradient at rest was 12.7 +/- 6.27 mm Hg and did not increase after exercise. Effective orifice areas were adequate, and the discharge coefficient ranged from 0.77 for the 21-mm prosthesis to 0.48 for the 29-mm prosthesis. Minimal regurgitation, which was washed out on the next systole, was observed with all sizes of the prosthesis. Disc opening was maximal (70 degrees) in all but one of the observed instances. Longer clinical follow-up is required, but the new integral monostrut Björk-Shiley prosthesis, with its important design changes and excellent hemodynamic performance, appears to be a promising aortic valve substitute.
D000328:Adult; D000368:Aged; D001021:Aortic Valve; D001022:Aortic Valve Insufficiency / Q000601:surgery*; D006328:Cardiac Catheterization; D005260:Female; D005500:Follow-Up Studies; D006350:Heart Valve Prosthesis / Q000592:standards*; D006439:Hemodynamics; D006801:Humans; D008297:Male; D008875:Middle Aged; D011474:Prosthesis Design; D013599:Systole; D013997:Time Factors