Double-beam flying spot scanner for two-dimensional polyacrylamide gel electrophoresis.
The construction of a double-beam photometer in which the light source is a cathode ray oscilloscope is described. The light spot from the oscilloscope was focused and reduced in size at the gel plane to give a diameter of less than 0.15 mm and make it possible to scan over a 50 X 59-mm rectangle; using reduced spatial resolution (spot less than 0.2 mm) the area scanned becomes 70 X 90 mm. The light from the CRT was divided into two beams; one was directed through the transparent object to a photomultiplier and the other to a reference photomultiplier. The signals from these two detectors were converted to the logarithm of the ratio by a logging amplifier to give a direct measure of absorbance. Positioning of the spot, control of light intensity, and measurement of absorbance were carried out through an interface to a 16-bit computer. The relationship between measured and actual absorbance was linear over the range of absorbance 0 to 2, which could be raised to 1 to 3 by placing a neutral filter in the reference beam. The system generated an image containing 256 X 256 pixels in about 5 min, the scanning speed was determined by the persistence time of the P4 phosphor on the cathode ray tube, and faster scans can be made using A6 phosphor.
D002138:Calibration; D003201:Computers; D004581:Electronics; D004591:Electrophoresis, Polyacrylamide Gel / Q000295:instrumentation*; D004867:Equipment Design; D055095:Optics and Photonics; D010783:Photometry / Q000295:instrumentation