Modified laminar flow biological safety cabinet.
Tests are reported on a modified laminar flow biological safety cabinet in which the return air plenum that conducts air from the work area to the high efficiency particulate air filters is under negative pressure. Freon gas released inside the cabinet could not be detected outside by a freon gas detection method capable of detecting 10(-6) cc/s. When T3 bacteriophage was aerosolized 5 cm outside the front opening in 11 tests, no phage could be detected inside the cabinet with the motor-filter unit in operation. An average of 2.8 x 10(5) plaque-forming units (PFU)/ft(3) (ca. 0.028 m(3)) were detected with the motor-filter unit not in operation, a penetration of 0.0%. Aerosolization 5 cm inside the cabinet yielded an average of 10 PFU/ft(3) outside the cabinet with the motor-filter unit in operation and an average of 4.1 x 10(5) PFU/ft(3) with the motor-filter unit not in operation, a penetration of 0.002%. These values are the same order of effectiveness as the positive-pressure laminar flow biological safety cabinets previously tested. The advantages of the negative-pressure return plenum design include: (i) assurance that if cracks or leaks develop in the plenum it will not lead to discharge of contaminated air into the laboratory; and (ii) the price is lower due to reduced manufacturing costs.
D000336:Aerosols; D000391:Air Microbiology; D000392:Air Movements; D000397:Air Pollution / Q000517:prevention & control*; D000399:Air Pressure; D005617:Chlorofluorocarbons, Methane; D003090:Coliphages / Q000302:isolation & purification; D004267:DNA Viruses; D004780:Environment, Controlled; D004864:Equipment and Supplies / Q000592:standards; D005374:Filtration; D007753:Laboratories
D000336:Aerosols; D005617:Chlorofluorocarbons, Methane