Sources of electron contamination for the Clinac-35 25-MV photon beam.
A detailed Monte Carlo approach has been employed to investigate the sources of electron contamination for the 25-MV photon beam generated by Varian's Clinac-35. Three sources of contamination were examined: (a) the flattening filter and beam monitor chamber, (b) the fixed primary collimators downstream from the monitor chamber and the adjustable photon jaws, and (c) the air volume separating the treatment head from the observation point. Five source-to-surface distances (SSDs) were considered for a single field size, 28 cm in diameter at 80 cm SSD. It was found that for small SSDs (80-100 cm), the dominant sources of electron contamination were the flattening filter and the beam monitor chamber which accounted for 70% of the unwanted electrons. Thirteen percent of the remaining electrons originated in the downstream primary collimators and the photon jaws, and 17% were produced in air. At larger SSDs, the fraction of unwanted electrons originating in air increased. At 400 cm SSD, 61% of the contaminating electrons present in the beam were produced in air, 34% originated in the flattening filter and beam monitor chamber, and 5% were due to interactions in the fixed collimators downstream from the monitor chamber and the adjustable photon jaws. These calculated results are substantiated by recent experiments.
D004583:Electrons; D009010:Monte Carlo Method; D010315:Particle Accelerators / Q000592:standards*; D011829:Radiation Dosage; D011882:Radiotherapy, High-Energy / Q000295:instrumentation*