Po2 temperature blood factor for blood gas apparatus.
PO2 temperature formulae supplied by manufacturers on automatic blood gas apparatus, PO2 corr. = PO2 37 degrees C X 10F X delta T were studied and compared to the experimental determination of the delta log PO2/delta T ratio (Hérigault et al. [10]). Acid-base status at 37 degrees C appeared to have a measurable influence on the PO2 temperature factor; alkalosis increased the delta log PO2/delta T ratio, and the contrary was found for acidosis in comparison with normal acid-base status at 37 degrees C. For the same PO2, measured at 37 degrees C, all the proposed formulae of commercial blood gas automatic apparatus did not give the same temperature corrected PO2. The observed difference between the corrected PO2 may be important and greater than the precision of the initial measurement. To correct the measured PO2 for temperature, a relationship between delta log PO2/delta T and PO2 is proposed, between PO2 zero and PO2 180 mmHg, which takes into account measured pH and PO2 values at 37 degrees C:delta log PO2/delta T = [(-0.35 pH + 0.658) X 10(-4) X PO2] + 0.035.
D000136:Acid-Base Equilibrium; D001784:Blood Gas Analysis / Q000379:methods*; D001831:Body Temperature; D006801:Humans; D008954:Models, Biological; D010100:Oxygen / Q000097:blood*; D010108:Oxyhemoglobins / Q000032:analysis; D012016:Reference Values
D010108:Oxyhemoglobins; D010100:Oxygen