Liquid and lyophilized quality-control materials compared for use in continuous-flow analysis.
Using a Technicon SMAC continuous-flow instrument, we compared a liquid quality-control material ("Decision level 3," Beckman) with two different lyophilized quality-control materials. The variances for 19 analytes in the three materials were significantly greater for Decision than for the Ortho Abnormal and the Omega II lyophilized control sera (p less than 0.00001 for each comparison by the omnibus test). The imprecision for Decision was not due to incomplete mixing or carryover. The viscosity of Decision, however, was more than twice that of the two lyophilized control sera or of pooled human serum.
D001322:Autoanalysis / Q000295:instrumentation / Q000592:standards*; D001774:Blood Chemical Analysis / Q000592:standards*; D006801:Humans; D011786:Quality Control; D013223:Statistics as Topic