[Application to dental casting machine of the rapid heating infrared image furnace (author's transl)].
The authors tried to manufacture a casting machine in dentistry by application of the infrared image furnace with a high heating speed and an easy control of a heating temperature. This machine melts an alloy in a carbon crucible set in the furnace, held in the horizontal position. Then, the furnace is turned to the vertical position to drop a melted alloy on the casting mold, and the alloy is cast in the mold by the pressure of Argon gas. The functions of trial casting machine were follows. 1. The trial casting machine was capable of heating to 1250 degrees C within one minute under 4 kW electric power. 2. The castability of the 20% Au-Pd-Ag commercial alloy cast in all casting conditions by the trial casting machine was higher than that of Thermotrol D-2 automatic centrifugal casting machine. 3. Castings of the trial casting machine showed higher tensile strength and elongation than those of the centrifugal casting machine, and the deviation of values got by the trial casting machine was small. In particular, some casting of the trial casting machine showed three times or over elongation values as compared with those of the centrifugal casting machine. 4. When casting conditions (casting temperature, casting pressure) of the trial casting machine changed, the physical properties of castings did not change so much. However, when the mold was not prevented from heating by the furnace in casting, the elongation of castings increased.
D003735:Dental Casting Technique / Q000295:instrumentation*; D006358:Hot Temperature; D007259:Infrared Rays; D008669:Metallurgy / Q000295:instrumentation