Direct withdrawal of zones during preparative capillary type isotachophoresis.
This study used a Shimadzu IP-1B capillary type isotachophoretic apparatus with a potential gradient detector. An ipp-1 withdrawal cell was fitted to this and a technique for withdrawing individual components directly through this port was developed using a microsyringe. The recovery rate was up to 45% for individual target components. When 100% withdrawal of the target component was attempted by withdrawing a volume four times the calculated volume (so that the zones both before and after the target component were also included), the best recovery rate was only 78%. In all cases, the results varied less than 3%. The limit for analysis of individual components of a 0.01 M solution was around 3 microliters. If this volume was exceeded, the ion quantity was too large for the volume of the microcapillary tube and mixed zones formed such that complete separation and analysis of individual components became impossible.
D002951:Citrates / Q000032:analysis; D019343:Citric Acid; D004586:Electrophoresis / Q000295:instrumentation* / Q000379:methods; D008094:Lithium / Q000032:analysis
D002951:Citrates; D019343:Citric Acid; D008094:Lithium