Performance of a refrigerated charcoal trap for xenon-133.
The impulse response function of a charcoal trap to a bolus of xenon-133 was determined as a function of the total number of hours run both at room temperature and at 25 degrees C. The peak of the response function for a new trap at room temperature reached a value of 360 MPC at 11 h. After 150 h of operation, the impulse response function was determined at -25 degrees C reaching a value of only 35 MPC at 25 h. The exhaust concentration of a trap in a busy nuclear medicine department using 150 mCi of xenon per week was measured and found to be 1600 MPC. The trap was placed in the freezer and kept there while it continued in use. Over a period of 3 weeks, the concentration of xenon in the exhaust of the trap dropped to a value of 13 MPC, or less than 1% of its value at room temperature.
D000398:Air Pollution, Radioactive / Q000517:prevention & control*; D002606:Charcoal; D006801:Humans; D012034:Refrigeration; D014980:Xenon Radioisotopes
D014980:Xenon Radioisotopes; D002606:Charcoal