Development of a reversible vas deferens occlusion device. VII. Physical and microscopic observations after long-term implantation of flexible prosthetic devices.
Flexible prosthetic devices developed previously to reversibly block sperm transport through the vas degerens were implanted in the vasa deferentia of dogs or kept in a semen bath for a 6-month period. Dimensional measurements, flow characteristics, leakage tests, and detent action force tests to displace the valve stem were performed with the valves. No changes occurred in any of these parameters with the exception of a decrease in the force required to change the position of the valve stem. This explains the occasional passage of spermatozoa through certain closed devices after they have been implanted for long periods of time. Histologic studies indicated that the same types of genital tract changes are associated with an implanted device as with vasectomy, and that they occur with approximately equal frequencies. Scanning electron microscope studies showed (1) excellent tissue ingrowth into the Dacron velour material of the implanted device; (2) the build-up of material, consisting in part of spermatozoa, in the device lumen and on the value stem; and (3) the absence of surface degradation of either the Silastic material or the valve stem of the device. It is concluded that no irreversible tissue damage is caused by the long-term implantation of flexible prosthetic devices into the vas deferens.
D000818:Animals; D003275:Contraceptive Devices, Male; D004285:Dogs; D004822:Epididymis / Q000033:anatomy & histology; D008297:Male; D008855:Microscopy, Electron, Scanning; D019736:Prostheses and Implants; D013083:Sperm Transport; D013091:Spermatogenesis; D014649:Vas Deferens / Q000033:anatomy & histology / Q000601:surgery*; D014659:Vasectomy / Q000009:adverse effects