title: Rapid, fully automated measurement of plasma homocyst(e)ine with the Abbott IMx analyzer.
abstract: We have developed a totally automated fluorescence polarization immunoassay for homocyst(e)ine with no pretreatment or chromatographic steps. Comparison with four well-established chromatographic methods yielded r values ranging from 0.980 to 0.997 and slopes from 1.030 to 1.493. Inter- and intraassay CVs ranged from 0.0% to 8.0% and from 0.0% to 6.4%, respectively. Imprecision (CV) of measuring six plasma samples on three instruments ranged from 6.3% to 10.2%. The assay was linear for plasma samples diluted with buffer from 0 to 8-fold. Mean recovery of homocysteine added to two plasma samples was 97.1% and 99.9%. The assay exhibited almost no cross-reactivity towards cysteine and methionine, and a batch of 20 samples can be processed in 60 min.
authors.affiliation: Abbott Diagnostics Division, Abbott Laboratories, Abbott Park, IL 60064, USA.
authors.forename: M T
authors.initials: MT
authors.lastname: Shipchandler
authors.forename: E G
authors.initials: EG
authors.lastname: Moore
mesh_terms: D001322:Autoanalysis / Q000379:methods*; D002138:Calibration; D002318:Cardiovascular Diseases / Q000097:blood; D016231:Fluorescence Polarization Immunoassay / Q000295:instrumentation / Q000379:methods* / Q000706:statistics & numerical data; D006710:Homocysteine / Q000097:blood*; D006801:Humans; D012307:Risk Factors; D012680:Sensitivity and Specificity
chemical_list: D006710:Homocysteine