Undesirable mode switching with a dual chamber rate responsive pacemaker.
The Telectronics 1250 Meta MV DDDR pacemaker is a new device featuring automatic mode switching from DDDR to VVIR pacing in the event of an atrial arrhythmia. Although mode switching is a valuable feature, sinus tachycardia can cause an undesirable mode switch to occur. Of 24 implants at this institution, 11 have been for an AV conduction disorder. Eight of these 11 patients were specifically evaluated for undesirable mode switching. During exercise testing and/or Holter monitoring, mode switching was repeatedly seen in seven of the eight at low levels of exercise. Factors precipitating mode switching were a low rate response factor, low upper rate setting, long base postventricular atrial refractory period (PVARP) and a long AV delay. During Holter monitoring, patients spent up to 50% of the time in VVIR pacing as opposed to DDDR pacing. It is concluded that patients with intact sinus node function are at risk of undesirable mode switching and should probably be programmed to the DDD mode unless there is a specific indication for DDDR pacing. If the DDDR mode is chosen, careful selection of the aforementioned pacing parameters is required.
Pitney
M R
MR
Eastern Heart Clinic, New South Wales, Australia.
May
C D
CD
Davis
M J
MJ
D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D004562:Electrocardiography; D005260:Female; D006339:Heart Rate; D006801:Humans; D008297:Male; D008875:Middle Aged; D010138:Pacemaker, Artificial; D005082:Physical Exertion; D012849:Sinoatrial Node / Q000503:physiopathology