title: Radio-frequency ring applicator: energy distributions measured in the CDRH phantom.
abstract: SAR distributions were measured in the CDRH phantom, a 1 cm fat-equivalent shell filled with an abdomen-equivalent liquid (sigma = 0.4-1.0 S m-1; dimensions 22 x 32 x 57 cm) to demonstrate the feasibility of the ring applicator to obtain deep heating. The ring electrodes were fixed in a PVC tube; diameter 48 cm, ring width 20 cm and gap width between both rings 31.6 cm. Radio-frequency energy was fed to the electrodes at eight points. The medium between the electrodes and the phantom was deionised water. The SAR distribution in the liquid tissue volume was obtained by a scanning E-field probe measuring the E-field in all three directions. With equal amplitude and phase applied to all feeding points, a uniform SAR distribution was measured in the central cross-section at 30 MHz. With RF energy supplied to only four adjacent feeding points (others were connected to a 50 omega load), the feasibility to perform amplitude steering was demonstrated; SAR values above 50% of the maximum SAR were measured in one quadrant only. SAR distributions obtained at 70 MHz showed an improved focusing ability; a maximum at the centre exists for an electric conductivity of the abdomen-equivalent tissue of 0.6 and 0.4 S m-1.
authors.affiliation: Department of Hyperthermia, Dr. Daniel den Hoed Cancer Center, Rotterdam, The Netherlands.
authors.forename: G C
authors.initials: GC
authors.lastname: van Rhoon
authors.forename: P
authors.initials: P
authors.lastname: Raskmark
authors.forename: S N
authors.initials: SN
authors.lastname: Hornsleth
authors.forename: P M
authors.initials: PM
authors.lastname: van den Berg
mesh_terms: D004574:Electromagnetic Fields; D006801:Humans; D006979:Hyperthermia, Induced / Q000295:instrumentation*; D008961:Models, Structural; D011846:Radio Waves