title: Can a drawover vaporizer be a pushover?
abstract: Bench testing was carried out to establish whether the vapour output from an OMV50 vaporizer, as used in the Triservice apparatus, differs according to whether the carrier gas is either drawn or pushed through the vaporizer. Results show that the differences in output concentration between the two modes were clinically insignificant.
authors: Army, Cambridge Military Hospital, Aldershot, Hampshire.
authors: J C
authors: JC
authors: Taylor
authors: J
authors: J
authors: Restall
mesh_terms: D000769:Anesthesia, Inhalation / Q000295:instrumentation*; D004305:Dose-Response Relationship, Drug; D006221:Halothane / Q000008:administration & dosage; D006801:Humans; D008887:Military Medicine / Q000295:instrumentation*; D009330:Nebulizers and Vaporizers; D011175:Positive-Pressure Respiration; D013990:Tidal Volume
chemical_list: D006221:Halothane