Biomechanical properties of small bone screws.
OBJECTIVE
To evaluate systematically the biomechanical properties of 13 popular screw designs, ranging from 0.8 to 2.0 mm in diameter.


METHODS
Screws were characterized in terms of external, core, and drill diameter; cutting flute and head design; material; pitch, thread depth; and height of shank (unthreaded portion) and shank with plate. They were tested in standardized bone specimens (2 x 2.5cm slabs of fresh bovine femur) 1, 2, 3, and 4-mm thick. For each screw-bone thickness combination, 10 trials were conducted to determine push-out force (POF) and another 10 trials to determine insertion (IT) and maximum torque (MT) yielding a total of 1,040 tests.


RESULTS
Among the 13 different screws, in 1-mm thick bone, both the lowest (108.5 N) and highest (294.9 N) POF were created by 2-mm screws (P < .001); that with the lowest POF had a long unthreaded shank and pitch, that with highest POF had a short unthreaded shank and pitch. Screws with 0.8- to 1.5-mm diameters showed no differences in POF. The 2-mm screw with the lowest POF also had the lowest MT in 1-mm thick bone compared with the other 2-mm screws (P < .001). In thicker bones (> 2 mm), two 2-mm screws showed 30% to 50% lower MT than the other same size screws (P < .001) because their head slots stripped easily. When all screws were considered together for a particular bone thickness, torque was strongly predicted by screw diameter (MT: r = .94, P < .001; IT: r = 0.92, P < .001). Screws with the same diameters varied significantly in IT because of differences in self-tapping cutting flute design.


CONCLUSIONS
External diameter, unthreaded shank height, head slot, and self-tapping cutting flute design had the greatest impact on screw strength and efficiency in thin cortical bone. Thread depth, core diameter, and metal type did not affect performance significantly. Under these highly standardized in vitro conditions, the ideal 2-mm screw has an unthreaded shank that is as short as possible, and the pitch is about 0.8 mm. Additional aspects of a clinical situation beside holding strength must, however, be considered when choosing a screw.
You
Z H
ZH
Department of Oral and Maxillofacial Surgery, Baylor College of Dentistry, Dallas, TX.
Bell
W H
WH
Schneiderman
E D
ED
Ashman
R B
RB
D000704:Analysis of Variance; D000818:Animals; D001696:Biomechanical Phenomena; D015519:Bone Density; D001863:Bone Screws; D002417:Cattle; D016009:Chi-Square Distribution; D004867:Equipment Design; D004868:Equipment Failure; D005269:Femur; D012044:Regression Analysis