Torsional and stiffness properties of Canal Master U stainless steel and nitinol instruments.
The purpose of this study was to compare stiffness and resistance to fracture of stainless steel and nickel titanium Canal Master U instruments. Instruments sizes 20 through 50 were tested according to ANSI/ADA Specification No. 28. Resistance to fracture was determined by twisting and measuring the maximum torque and angular deflection (rotation) at the failure point. Stiffness was determined by measuring the moment required to bend the instrument 45 degrees. Stainless steel and nickel titanium Canal Master U instruments satisfied and far exceeded specification standards for stiffness. They also satisfied and far exceeded the standards for angular deflection at the failure point. Stainless steel Canal Master U met or exceeded the maximum torque at failure standards in all sizes except 50, whereas the nickel titanium instruments met the standards in all sizes except 45 and 50. The mechanical properties of nickel titanium, especially its low bending moment and high angular rotation at failure, encourage further evaluation with clinical studies.
Camps
J
J
Department of Restorative Dentistry and Endodontics, Faculty of Dentistry, Unité IMEB, Marseilles, France.
Pertot
W J
WJ
D000497:Alloys / Q000737:chemistry*; D000704:Analysis of Variance; D003762:Dental Instruments; D003799:Dental Stress Analysis; D004548:Elasticity; D004868:Equipment Failure; D008422:Materials Testing; D009532:Nickel / Q000737:chemistry; D018583:Pliability; D012390:Root Canal Therapy / Q000295:instrumentation*; D012399:Rotation; D013193:Stainless Steel / Q000737:chemistry*; D013718:Tensile Strength; D014025:Titanium / Q000737:chemistry*
D000497:Alloys; D013193:Stainless Steel; C013616:nitinol; D009532:Nickel; D014025:Titanium