Pediatric defibrillation: current flow is improved by using "adult" electrode paddles.
OBJECTIVE
Current flow, the major determinant of defibrillation success, depends on delivered energy and transthoracic impedance. Previous experimental data suggest that transthoracic impedance is higher using "pediatric" electrode paddles compared to the larger "adult" electrode paddles. There are few data from actual shocks to support the experimental studies. The purpose of this study was to measure transthoracic impedance during actual shock delivery and to determine the optimal electrode paddle size for pediatric defibrillation and cardioversion.


METHODS
We prospectively evaluated all shocks of < or = 20 joules given to pediatric patients at the University of Iowa from 1988 to 1992. Data collected included energy selected by the operator, energy delivered by the defibrillator, peak current flow and transthoracic impedance. Data were analyzed by unpaired t test and linear regression.


RESULTS
Fifty-five shocks were delivered to 20 patients, age newborn to 8 years. Thirty-seven shocks were given with "pediatric" electrode paddles (surface area 21 cm2) and 18 with "adult" electrode paddles (surface area 83 cm2). Selected energy correlated well with delivered energy (r = .98, P < .0001). Delivered energy using "pediatric" electrodes did not differ from that delivered with "adult" electrodes (8.0 +/- 0.9 joules vs 10.5 +/- 1.2 joules P > .1). However transthoracic impedance was significantly higher with "pediatric" electrodes (78.1 +/- 4.4 ohms vs 54.6 +/- 2.7 ohms P < .0008), resulting in lower peak current flow through "pediatric" electrode paddles (6.2 +/- 0.5 amps vs 8.7 +/- 0.5 amps P < .002). There was no correlation between joules/kg and peak current flow (r = .26, P > .05).


CONCLUSIONS
Use of "pediatric" electrode paddles results in higher transthoracic impedance and thus lower peak current flow. In pediatric defibrillation, larger "adult" electrode paddles should be used as soon as chest size permits (approximately 10 kg). Lower transthoracic impedance results in higher current flow that facilitates cardioversion and defibrillation.
Atkins
D L
DL
Department of Pediatrics, University of Iowa, Iowa City 52242.
Kerber
R E
RE
D002648:Child; D002675:Child, Preschool; D004554:Electric Countershock / Q000295:instrumentation*; D017097:Electric Impedance; D004560:Electricity; D004566:Electrodes; D006801:Humans; D007223:Infant; D007231:Infant, Newborn; D011446:Prospective Studies