Automated ventilator testing.
A new era has arrived for the Biomedical Engineering Department at the Royal Women's Hospital in Melbourne. We have developed a system to qualitatively test for intermittent or unconfirmed faults, associated with Bear Cub ventilators. Where previous testing has been inadequate, computer logging is now used to interface the RT200 Timeter Calibration Analyser (TCA) to obtain a real time display of data, which can be stored and graphed. Using Quick Basic version 4.5, it was possible to establish communication between the TCA and an IBM compatible computer, such that meaningful displays of machine performance were produced. From the parameters measured it has been possible to obtain data on Peak Pressure, Inspiratory to Expiratory ratio (I:E ratio) Peak Flow and Rate. Monitoring is not limited to these parameters, though these were selected for our particular needs. These parameters are plotted in two ways: 1. Compressed average versus time, up to 24 hours on one screen 2. Raw data, 36 minutes displayed on each screen. The compressed data gives an overview which allows easy identification of intermittent faults. The uncompressed data confirms that the averaged signal is a realistic representation of the situation. One of the major benefits of this type of data analysis, is that ventilator performance may be monitored over a long period of time without requiring the presence of a service technician. It also allows individual ventilator performance to be graphically compared to other ventilators.
Ghaly
J
J
Biomedical Engineering Department, Royal Women's Hospital, Melbourne.
Smith
A L
AL
D001315:Australia; D001698:Biomedical Engineering / Q000295:instrumentation* / Q000379:methods; D002138:Calibration; D001330:Electronic Data Processing; D004868:Equipment Failure; D008838:Microcomputers; D012122:Ventilators, Mechanical