The Harris-Galante cementless acetabular component: results in 190 cases with at least 3 years follow-up.
The results of 190 primary total hip arthroplasties with a Harris-Galante cementless acetabular cup were reviewed. All patients had a minimum follow-up of 3 years (range, 3 to 5.5 years, mean 46 months). Clinical and radiographical analysis was performed. Inguinal pain was recorded in five cases. We noted a fracture of a screw in four cases without further implications. There was no evidence of acetabular loosening. There was no migration of the acetabular cup. No acetabular component showed measurable wear of the polyethylene liner. Non-progressive radiolucent lines were recorded in 14% of the patients: among these patients, radiolucent lines were noted in zone 1 in 46%, in zone 2 in 4% and in zone 3 in 86%. Two socket revisions became necessary. One patient suffered a deep-seated infection. Another revision was necessary because of recurrent dislocation.
Claus
B
B
Aalsters Stedelijk Ziekenhuis, Aalst, Belgium.
Van Innis
B
B
De Witte
E
E
Van Overschelde
J
J
Magotteaux
B
B
Fatemi
F
F
Vandepaer
F
F
D000077:Acetabulum / Q000601:surgery; D000328:Adult; D000368:Aged; D000369:Aged, 80 and over; D005500:Follow-Up Studies; D006622:Hip Prosthesis; D006801:Humans; D008875:Middle Aged; D011474:Prosthesis Design