[Computer-assisted measurement of ocular misalignment in infants and young children using the digital Purkinje reflection pattern procedure].
A digital image recording and processing system is presented that allows a quick diagnosis of microstrabismus in non-cooperative children. It is thus particularly suited for screening purposes.


METHODS
The Purkinje Reflection Pattern Evaluation (RPE) method is used: three small flashes are used to produce the desired Purkinje images. Two horizontal rows of the three 1st Purkinje images (anterior corneal reflections) and of the three 4th Purkinje images (posterior crystalline lens reflections) stemming from the three light sources form the characteristic Purkinje image reflection pattern. Each eye's position is calculated from the shift between the upper and lower rows of reflections by means of two simple formulae. From the angles obtained in binocular fixation and monocular fixation the manifest angle of strabismus corresponding to the angle measured in the simultaneous prism-and-cover test is computed. The measurement is performed at a fixation distance of 50 cm under natural viewing conditions. To obtain a picture one only has to get the child's attention for a short moment. The primary position is triggered with the fixation light, which is operated by a switch.
The digital image recording is done with a hand-held device comprising two miniaturized video cameras, three photo flashes and a fixation light that is operated manually. An IBM-compatible PC equipped with a hard disk and two frame grabbers was adapted for the storage and processing of the pictures. The pictures are evaluated interactively in a few minutes on the workstation's monitor immediately after the measurement. To this end specially designed menu-driven software was implemented.


RESULTS
Examples of the measuring procedure and clinical results in infants with microtropic highlight the potential of the system as a screening apparatus and for the exact measurement of small and large squint angles. Usually even 1-year-old children can cooperate well enough to get good-quality pictures in binocular fixation. The new digital system allows easy and rapid application of the Purkinje Reflection Pattern Evaluation method since the time-consuming photographic film processing and evaluation are no longer necessary. For the first time small angles of strabismus under 5 degrees (10 PD) can be measured with a precision of less than 1 degree (2 PD) under clinical conditions in non-cooperative children.
Barry
J C
JC
Augenklinik, Medizinische Einrichtungen, RWTH Aachen.
Effert
R
R
Kaupp
A
A
Kleine
M
M
Reim
M
M
D000328:Adult; D002648:Child; D002675:Child, Preschool; D003315:Cornea / Q000503:physiopathology; D005260:Female; D005403:Fixation, Ocular / Q000502:physiology*; D006801:Humans; D007091:Image Processing, Computer-Assisted / Q000295:instrumentation*; D007223:Infant; D007908:Lens, Crystalline / Q000503:physiopathology; D008297:Male; D010775:Photic Stimulation / Q000295:instrumentation*; D010781:Photography / Q000295:instrumentation*; D012016:Reference Values; D013285:Strabismus / Q000175:diagnosis* / Q000235:genetics / Q000503:physiopathology; D014741:Video Recording / Q000295:instrumentation
Computergestützte Messung von Augenfehlstellungen bei Säuglingen und Kleinkindern mit Hilfe des digitalen Purkinje-Reflexmuster-Verfahrens.