[Digital processing and evaluation of ultrasound images].
With the help of workstations and PCs, on-site image processing has become possible. If the images are not available in digital form the video signal has to be A/D converted. In the case of colour images the colour channels R (red), G (green) and B (blue) have to be digitized separately. "Truecolour" imaging calls for an 8 bit resolution per channel, leading to 24 bits per pixel. Out of a pool of 2(24) possible values only the relevant 128 gray values and 64 shades of red and blue respectively needed for a colour-coded ultrasound image have to be isolated. Digital images can be changed and evaluated with the help of readily available image evaluation programmes. It is mandatory that during image manipulation the gray scale and colour pixels and LUTs (Look-Up-Table) must be worked on separately. Using relatively simple LUT manipulations astonishing image improvements are possible. Application of simple mathematical operations can lead to completely new clinical results. For example, by subtracting two consecutive colour flow images in time and special LUT operations, local acceleration of blood flow can be visualized (Colour Acceleration Imaging).
Borchers
J
J
Philips Medizin Systeme, Hamburg.
Klews
P M
PM
D003116:Color; D003199:Computer Systems; D006801:Humans; D007091:Image Processing, Computer-Assisted / Q000295:instrumentation*; D014463:Ultrasonography / Q000295:instrumentation*; D014741:Video Recording / Q000295:instrumentation
Digitale Verarbeitung und Auswertung von Ultraschallbildern.