The additional work of breathing imposed by Mapleson A systems.
The additional work attributable to breathing through five Mapleson A anaesthetic breathing systems (Magill, Lack, Parallel Lack, Humphrey ADE and Enclosed Magill) was studied using a lung model. With all five systems, the additional work was found to be a function of fresh gas flow, respiratory flow as well as system geometry. Within the range of fresh gas flow and respiratory flow studied, the additional work ranged between 80 mJ.l-1 and 182 mJ.l-1. Expiratory work was always greater than the inspiratory workload. Increasing fresh gas inflow into the system increases expiratory work, both resistive and elastic components. The Magill system posed the least work expenditure. The values for the additional work obtained with the lung model were of the same order of magnitude when measurements were taken in volunteers.
Ooi
R
R
Magill Department of Anaesthetics, Westminster Hospital, London.
Pattison
J
J
Soni
N
N
D000328:Adult; D000769:Anesthesia, Inhalation / Q000295:instrumentation*; D005260:Female; D006801:Humans; D008168:Lung / Q000502:physiology; D008297:Male; D008961:Models, Structural; D010100:Oxygen / Q000008:administration & dosage; D012123:Pulmonary Ventilation / Q000502:physiology; D012119:Respiration / Q000502:physiology*; D014939:Work of Breathing / Q000502:physiology*
D010100:Oxygen