Malfunction of medical equipment as a result of mains borne interference.
Medical equipment has become more intelligent as the manufacturers have incorporated the latest microprocessor based technology. Equipment malfunction can be caused at any time by inherent errors in the control program but it is particularly important that this is designed to cope with the effects of electrical interference which, in addition, may cause corruption of the software. We have considered interference found in the mains supply in the hospital environment. Using a test protocol with appropriate interference simulators, a wide range of medical equipment was removed temporarily from use and its immunity to electrical mains borne interference tested. Battery operated mains rechargeable devices were unaffected by mains voltage variations including drop-outs and sags whereas mains powered devices were affected to varying degrees of severity. In particular, repetitive drop-outs caused loss of power due to fuse blowing in some life support equipment. Impulses affected 25% and pulse bursts 50% of the equipment tested with some evidence that the more recent designs coped better. The EEC Directive on electro-medical compatibility compliance may cause the design of equipment to be improved but hospitals will have to cope with the above problems in their existing equipment for many years to come.
Railton
R
R
Monklands Hospital, Airdrie, UK.
Currie
G D
GD
Corner
G A
GA
Evans
A L
AL
D004560:Electricity; D004868:Equipment Failure / Q000706:statistics & numerical data*; D004865:Equipment and Supplies, Hospital; D006801:Humans; D012606:Scotland