Electromagnetic pulse (EMP), Part II: Field-expedient ways to minimize its effects on field medical treatment facilities.
Part I of this paper showed that a field commander can expect approximately 65% of his unprotected electronic medical equipment to be damaged by the electromagnetic pulse (EMP) from a single nuclear detonation as far as 2200 km away. Using computer modeling, field-expedient ways to minimize the effects of EMP were studied. The results were: (1) keep wiring near the ground, (2) keep wiring short, (3) unplug unused equipment, (4) run power cabling and tents in a magnetic north-south direction (avoid running power cabling in the east-west direction), and (5) place sensitive equipment in International Organization for Standardization shelters.
Vandre
R H
RH
USA Institute of Dental Research, Walter Reed Army Medical Center, Washington, DC 20307-5300.
Klebers
J
J
Tesche
F M
FM
Blanchard
J P
JP
D003198:Computer Simulation; D004582:Electronics, Medical / Q000295:instrumentation*; D006801:Humans; D008280:Magnetics; D008887:Military Medicine / Q000295:instrumentation; D009689:Nuclear Warfare; D011827:Radiation; D014481:United States