A survey of the acoustic outputs of diagnostic ultrasound equipment in current clinical use.
Surveys published up to 1991 have highlighted a steady increase in the acoustic outputs from diagnostic ultrasound equipment. Since 1991 we have made measurements of the maximum peak negative pressure (p-) and spatial peak temporal average intensity (ISPTA) produced by 223 probes from 82 scanning systems in current clinical use in the Northern Region in the UK. Measurements have also been made of the maximum total acoustic power generated by 45 probes from 17 scanners. The results from these measurements are presented in this article and compared to the results of a similar survey of equipment from both the Northern and Wessex Regions in the UK and published in 1991. The comparison shows that measured ISPTA values have increased approximately sixfold in B mode and approximately threefold in colour Doppler mode. Also, measured total acoustic power values have doubled in pulsed Doppler mode. The present survey also draws attention to some particularly high ISPTA values obtained from a number of probes and scanning systems. This survey has shown that measurements of acoustic outputs from diagnostic ultrasound scanners in current clinical use are substantially higher than reported in earlier surveys and, for certain scanners, the acoustic outputs from scanned beam modes of operation can reach levels hitherto only found in pulsed Doppler mode.
Henderson
J
J
Regional Medical Physics Department, Newcastle General Hospital, Newcastle Upon Tyne, UK.
Willson
K
K
Jago
J R
JR
Whittingham
T A
TA
D000162:Acoustics; D002138:Calibration; D003625:Data Collection; D004869:Equipment Safety; D006801:Humans; D013677:Medical Laboratory Science / Q000592:standards*; D014463:Ultrasonography / Q000295:instrumentation / Q000592:standards*