Force delivery of Ni-Ti coil springs.
Sentalloy springs (GAC, Central Islip, N.Y.) of the open and closed type were investigated with a special designed device. The closed coil springs were subjected to a tensile and the open coil springs to a compression test. After a first measurement, the springs were activated for a period of 4 weeks and then reinvestigated with the same procedure. It could be shown distinctly that, with the different coil springs, the force delivery given by the producer could be achieved only within certain limits. To remain in the martensitic plateau, changed activation ranges, and for the Sentalloy coil springs white and red of the open and closed type, also changed force deliveries had to be taken into account. There was a distinct decrease in force delivery between the first and second measurement. After considering the loading curves of all the Sentalloy coil springs and choosing the right activation range respective to the force delivery, it was found that the coil springs deliver a superior clinical behavior and open new treatment possibilities.
Manhartsberger
C
C
Department of Orthodontics, University Clinic for Dentistry, Innsbruck, Austria.
Seidenbusch
W
W
D003481:Cuspid; D003722:Dental Alloys; D003799:Dental Stress Analysis; D004548:Elasticity; D006801:Humans; D008422:Materials Testing; D009532:Nickel; D009967:Orthodontic Appliances; D018709:Statistics, Nonparametric; D014025:Titanium; D014087:Tooth Movement Techniques / Q000295:instrumentation; D014095:Tooth, Impacted / Q000628:therapy
D003722:Dental Alloys; C040654:titanium nickelide; D009532:Nickel; D014025:Titanium