Prospective, randomized comparison of the Flotem Iie and Hotline fluid warmers in anesthetized adults.
OBJECTIVE
To compare the fluid warming capabilities of the Hotline and Flotem IIe devices in surgical patients, and whether warming intravenous (i.v.) fluids with the Hotline device resulted in less hypothermia and less need for other warming methods compared with the Flotem IIe device.


METHODS
Part 1. Prospective, nonrandomized. Part 2. Prospective, randomized.


METHODS
Teaching hospital, tertiary care center.


METHODS
Part 1. 24 adult patients undergoing elective surgery. Part 2. 49 adult patients scheduled for major elective orthopedic or gynecologic surgery of greater than 2 hours' duration with general anesthesia.


METHODS
Part 1. Insertion of sterile in-line thermistors along the path of fluids delivered using the Hotline or Flotem device. Part 2. Patients were randomly assigned to receive i.v. fluids via the Hotline (n = 21) or Flotem IIe (n = 18) warmers. Intervention for core hypothermia [lower esophageal temperature (Teso) less than 35.5 degrees C], was with forced air warming. Intervention for postoperative shivering was with meperidine by a nurse who was blinded to the treatment group.


METHODS
Part 1. Temperature of infused fluids before and after the warmer (T in and T out) and before the fluid entered the patient after insertion of a standard 84 cm extension set having a stopcock for injection of medications (T distal). Part 2. Tympanic (Ttym) was recorded before induction of anesthesia (baseline) and at 15-minute intervals after induction. Teso was recorded at 15-minute intervals after induction. The maximum intraoperative Ttym decline from baseline (delta Ttym max) was calculated.


RESULTS
Part 1. Flow rates were between 1 and 33 ml/min. T out and T distal (mean +/- SEM) of fluids infused through the Hotline device were warmer compared with those infused via the Flotem IIe device (Hotline T out: 35.7 +/- 0.1 degrees C and Hotline T distal: 33.4 +/- 0.2 degrees C vs. Flotem IIe T out: 28.9 +/- 0.2 degrees C and Flotem IIe T distal: 28.3 +/- 0.2 degrees C; p < 0.001 between warmers). Part 2. Maximal decrease in Ttym from preoperative baseline (before intervention) was greater in the Flotem IIe compared with the Hotline group (delta Ttym max = -1.4 +/- 0.1 vs. -0.9 +/- 0.1 degree C, p = 0.01). Five patients in the Flotem IIe group required forced air warming for treatment of hypothermia versus none in the Hotline group (p < 0.01). Postoperatively, five patients in the Flotem IIe group required treatment with meperidine for severe shivering versus one patient in the Hotline group (p < 0.05).


CONCLUSIONS
The Hotline device delivered fluids to the patient at consistently warmer temperatures compared with the Flotem IIe device during actual clinical conditions. This was associated with maintenance of near normal core temperatures throughout the procedure in the Hotline group, and a decreased need for interventions such as forced-air warming and treatment for severe shivering.
Patel
N
N
Department of Anesthesiology, MetroHealth Medical Center, Cleveland, OH 44109, USA.
Smith
C E
CE
Pinchak
A C
AC
Hagen
J F
JF
D000328:Adult; D000776:Anesthesiology / Q000295:instrumentation*; D001831:Body Temperature / Q000502:physiology*; D005069:Evaluation Studies as Topic; D005260:Female; D006801:Humans; D007432:Intraoperative Period; D008297:Male; D008875:Middle Aged; D011446:Prospective Studies