A new beaded carbon molecular sieve sorbent for 222Rn monitoring.
A new commercially available beaded carbon molecular sieve sorbent, Carboxen-564 (20/45 mesh), was tested and compared to Calgon-PCB (40/80) activated carbon for its adsorptive and desorptive characteristics under controlled conditions of temperature (25 degrees C) and relative humidity (RH). The amount of water vapor adsorbed by the beaded carbon molecular sieve material was typically a factor of 4 lower than the activated carbon, with a concomitant fourfold increase in the 222Rn adsorption coefficient, K(Rn). The maximum K(Rn) value for a thin layer of Carboxen-564, following a 2-d exposure at 40% RH, was 7.2 Bq kg(-1) per Bq m(-3). The K(Rn) or a 1-cm bed, following a 2-d exposure was 5.5 Bq m(-3), a 25% reduction. Under dynamic sampling conditions, where 0.4 g of the beaded carbon molecular sieve was contained in a 6 cm x 0.4 cm diameter tube, the maximum K(Rn) value was 6.5 Bq m(-3) after 2.5 h of sampling at 29% RH when the input flow rate was 4.2 x 10(-3) m3 h-1. Kinetic studies were also conducted under passive sampling conditions. The data show that the 222Rn buildup time-constant for a thin layer of the beaded carbon molecular sieve material was 1.3 h, whereas that of a 1 cm bed was 13 h. The 222Rn desorption time-constants, from gram amounts of the beaded carbon molecular sieve material into air and into a commercially available toluene based liquid scintillation cocktail, were 2 h and 3 h, respectively. Carboxen's high 222Rn adsorbing capacity, rapid kinetics, hydrophobicity and physical properties makes it an attractive alternative to other commercially available activated carbon used in passive and dynamic sampling devices.
Scarpitta
S C
SC
Environmental Measurements Laboratory, U.S. Department of Energy, New York 10014-3621, USA.
D000327:Adsorption; D011834:Radiation Monitoring / Q000379:methods*; D011886:Radon / Q000032:analysis*
D011886:Radon