Clamping stiffness and its influence on load distribution between paired internal spinal fixation devices.
The load distribution between two internal spinal fixation devices depends, besides other factors, on their stiffness. The stiffness ranges were determined experimentally for the clamps of the AO internal fixator with lateral nut and with posterior nut as well as for the clamps of the SOCON fixator. The stiffness of eight devices each differed by a factor of 3.1 for the clamp with lateral nut, by a factor of 1.5 for the clamp with posterior nut, and by a factor of 1.4 for the clamp of the SOCON fixator. For the AO clamp with lateral nut, the influence of the nut-tightening torque on the stiffness was determined. Using instrumented internal spinal fixation devices mounted to plastic vertebrae and simulating a corpectomy, the load distribution between the implants was measured for different tightening torques. It could be shown that, for the AO internal fixator whose clamps have a lateral nut, a nut-tightening torque > 5 Nm has only a negligible influence on load-sharing between the implants. Tooth damage occurs when the teeth of the clamp body and clamping jaw of the clamp with lateral nut do not gear together exactly, which leads to changes in the clamping stiffness and load-sharing between the two implants.
Rohlmann
A
A
Department of Biomechanics, Free University of Berlin, Oskar-Helene-Heim, Germany.
Calisse
J
J
Bergmann
G
G
Radvan
J
J
Mayer
H M
HM
D006801:Humans; D016268:Internal Fixators; D018583:Pliability; D015203:Reproducibility of Results; D013131:Spine; D013718:Tensile Strength