Two cases of fractured esophageal nitinol stents.
We report here on two cases of fractured nitinol stents in the esophagus. In case 1, the correctly inserted stent broke spontaneously shortly after insertion. In a second case, a nitinol stent broke after laser application due to tumor ingrowth with massive bleeding. In both cases, a second stent was implanted in order to reestablish food passage. The fracture of the stent in case 1 seemed to be caused by defective material, whilst in case 2 the stent broke because of thermal overstrain during laser application. The use of electrocoagulation or laser in the stent area should therefore be avoided; argon plasma coagulation may offer an effective alternative in treating tumor ingrowth. As there was a risk from piercing broken filaments with the second stent, covering a fracture using stents with tight walls or plastic tubes seems to be a more effective approach than the inserting an uncovered stent type.
Schoefl
R
R
Dept. of Gastroenterology and Hepatology, University of Vienna, Austria.
Winkelbauer
F
F
Haefner
M
M
Poetzi
R
R
Gangl
A
A
Lammer
J
J
D000368:Aged; D000497:Alloys; D002294:Carcinoma, Squamous Cell / Q000628:therapy*; D003251:Constriction, Pathologic; D004938:Esophageal Neoplasms / Q000628:therapy*; D017809:Fatal Outcome; D005260:Female; D006801:Humans; D015607:Stents
D000497:Alloys; C013616:nitinol