title: Measurement of ionized magnesium with AVL 988/4 electrolyte analyzer: preliminary analytical and clinical results.
abstract: Only free magnesium has biological activity: technology for measuring the ionized fraction of magnesium is now available via ion-selective electrodes. We have evaluated an instrument (AVL 988/4) which determines ionized magnesium (cMg2+) with an ion-selective electrode based on the ionophore ETH 7025. The selectivity of the electrode is adequate for the ions normally present in plasma, except for calcium: the interference is automatically corrected by simultaneous measurements of calcium with compensation for the calcium interference to the magnesium signal. We have first verified possible interference caused by sampling procedures: known silicon interference has been avoided by use of glass tubes (BD Vacutainer with no additive, code 7626); heparin interference has been measured and found significant above 20 UI.mL-1 of plasma. Instrument evaluation according to NCCLS protocol gives the following imprecision results on 20 replicated analyses: cMg2+ (mmol.L-1) 1.29, 0.76, 0.23, CVs% (within-run) 0.67, 0.67, 3.00 and CVs% (between-run) 4.06, 3.91, 5.89 respectively. Linearity (in the range 0.23-1.60 mmol.L-1) was: measured cMg2+ = 0.981.(calculated cMg2+) + 0.009 mmol/L; r = 0.999. In healthy adults (n = 103) cMg2+ was in the range 0.46-0.74 mmol.L-1 (with a mean of 0.60 mmol/L and normal distribution). These values represent 57% to 84% of serum total magnesium concentration (TMg) (mean 71%). pH dependence of cMg2+ is present, usually to a lower extent with respect to cCa2+, but it seems different in patients with real or in vitro provoked acidosis and in hemodialyzed patients. Citrate interference on ionized magnesium measurements was found both in vitro and in vivo, whilst that due to lactate was demonstrated only in vitro. On a wide range of cMg2+ (n = 100), a good correlation is obtained both with TMg and ultrafiltrable Mg (UFMg): cMg2+ = 0.723.TMg + 0.008 mmol.L-1, r = 0.978; cMg2+ = 0.912.UFMg + 0.10 mmol.L-1, r = 0.968, respectively. The ionized magnesium in ultrafiltrate was found 25% lower than that in serum. The lifespan of the electrode, evaluated on the basis of both time from installation and on number of measured samples, was estimated longer than 4 months and able to analyze more than 1500 samples, whichever comes first. The four electrodes we used during 18 months behaved all the same way. The correlation between measurements performed in whole blood (WB-cMg2+) and in the corresponding serum (S-cMg2+) was excellent: WB-cMg2+ = 0.954.S-cMg2+ +0.02 mmol.L-1; r = 0.998; n = 60.
authors.affiliation: Clinical Biochemistry and Haematology Laboratory, Niguarda Cà Granda Hospital, Milan, Italy.
authors.forename: F
authors.initials: F
authors.lastname: Zoppi
authors.forename: A
authors.initials: A
authors.lastname: De Gasperi
authors.forename: E
authors.initials: E
authors.lastname: Guagnellini
authors.forename: A
authors.initials: A
authors.lastname: Marocchi
authors.forename: E
authors.initials: E
authors.lastname: Mineo
authors.forename: F
authors.initials: F
authors.lastname: Pazzucconi
authors.forename: C
authors.initials: C
authors.lastname: Rossi
authors.forename: D
authors.initials: D
authors.lastname: Turrini
mesh_terms: D000328:Adult; D019343:Citric Acid; D004573:Electrolytes / Q000032:analysis / Q000097:blood; D006493:Heparin / Q000378:metabolism; D006801:Humans; D006863:Hydrogen-Ion Concentration; D017736:Ion-Selective Electrodes; D007477:Ions; D019344:Lactic Acid; D016014:Linear Models; D008274:Magnesium / Q000032:analysis* / Q000097:blood*; D013677:Medical Laboratory Science / Q000295:instrumentation*; D011199:Potentiometry / Q000295:instrumentation; D006435:Renal Dialysis; D015203:Reproducibility of Results; D012995:Solubility; D013048:Specimen Handling; D014462:Ultrafiltration
chemical_list: D004573:Electrolytes; D007477:Ions; D019343:Citric Acid; D019344:Lactic Acid; D006493:Heparin; D008274:Magnesium