[Evaluation of vaporizers by anesthetic gas monitors corrected with a new method for preparation of calibration gases].
For resolving the discrepancy of concentrations found among anesthetic gas monitors, the author proposed a new method using a vaporizer as a standard anesthetic gas generator for calibration. In this method, the carrier gas volume is measured by a mass flow meter (SEF-510 + FI-101) installed before the inlet of the vaporizer. The vaporized weight of volatile anesthetic agent is simultaneously measured by an electronic force balance (E12000S), on which the vaporizer is placed directly. The molar percent of the anesthetic is calculated using these data and is transformed into the volume percent. These gases discharging from the vaporizer are utilized for calibrating anesthetic gas monitors. These monitors are normalized by the linear equation describing the relationship between concentrations of calibration gases and readings of the anesthetic gas monitors. By using normalized monitors, flow rate-concentration performance curves of several anesthetic vaporizers were obtained. The author concludes that this method can serve as a standard in evaluating anesthetic vaporizers.
Kurashiki
T
T
Department of Anesthesiology, San-in Rosai Hospital, Yonago.
D018685:Anesthetics, Inhalation; D002138:Calibration; D004784:Environmental Monitoring / Q000295:instrumentation*; D005069:Evaluation Studies as Topic; D009330:Nebulizers and Vaporizers / Q000592:standards*
D018685:Anesthetics, Inhalation