title: The multileaf collimator for fast neutron therapy at Louvain-la-Neuve.
abstract: The multileaf collimator of the fast neutron therapy facility at Louvain-la-Neuve is described, as well as some of the physics experiments performed in order to evaluate the attenuation of neutron beams in different materials and thus optimize the composition of the collimator leaves. The multileaf collimator consists of two sets of 22 leaves each, which can be moved independently. They are made of iron and their thickness is 95 cm. Seven borated polyethylene disks are located in the distal part of the leaves in order to absorb more efficiently the low-energy component of the neutron spectrum. The width of the leaves is 1 cm at their distal part. The leaves can move 11 cm outwards and 6 cm inwards from their reference position, and field sizes up to 25.7 x 24.8 cm, as well as irregular field shapes, can be obtained. The inner part of the leaves and their two sides are always focused on the target. The complete multileaf collimator can rotate around the beam axis, from -90 degrees to +90 degrees from the reference position. The width of the penumbra (80-20% isodoses) is 0.64 cm and 1.17 cm at the depth of the maximum buildup and at 10 cm in depth respectively, for a 10 x 10 cm field size. The collimator is adequate for the energy of the p(65)+Be neutron beam of Louvain-la-Neuve and has been adapted to the fixed vertical beam. It has been designed following the original plans of Scanditronix, adjusted and fully assembled at the workshop of the Centre de Recherches du Cyclotron (CRC). Systematic measurements were performed in order to optimize the design and the composition of the leaves. In particular the attenuations of the actual beam and of monoenergetic neutron beams were measured in different materials such as iron and polyethylene. Above (upstream) the multileaf collimator, a fixed pre-collimator (iron thickness 50 cm; section 1 x 1 m) defines a conical aperture aligned on the largest opening of the leaves. It contains the two transmission chambers and a 2 cm thick polyethylene layer used for hardening the beam.
authors.affiliation: Service de Radiothérapie, Cliniques Universitaires St-Luc, Brussels, Belgium.
authors.forename: J M
authors.initials: JM
authors.lastname: Denis
authors.forename: J P
authors.initials: JP
authors.lastname: Meulders
authors.forename: E
authors.initials: E
authors.lastname: Lannoye
authors.forename: Y
authors.initials: Y
authors.lastname: Longrée
authors.forename: G
authors.initials: G
authors.lastname: Ryckewaert
authors.forename: F
authors.initials: F
authors.lastname: Richard
authors.forename: S
authors.initials: S
authors.lastname: Vynckier
authors.forename: A
authors.initials: A
authors.lastname: Wambersie
mesh_terms: D001530:Belgium; D001608:Beryllium; D002173:Cancer Care Facilities; D017357:Cyclotrons; D004867:Equipment Design; D005214:Fast Neutrons; D006801:Humans; D011879:Radiotherapy Dosage; D011882:Radiotherapy, High-Energy / Q000295:instrumentation / Q000379:methods
chemical_list: D001608:Beryllium