title: Occlusion of congenital ventricular septal defects by the buttoned device. "Buttoned device" Clinical Trials International Register.
abstract: OBJECTIVE
  To study the feasibility of congenital ventricular septal defect occlusion by the buttoned device and to establish guidelines for its safe and effective application.
  
  
  METHODS
  A descriptive study of all patients with a congenital ventricular septal defect undergoing transcatheter occlusion with the buttoned device, from March 1994 to May 1995. These patients were otherwise candidates for elective surgery at their institutions because they had persistence of a significant shunt (Qp:Qs = 1.5-2.1:1, median = 1.7), with left ventricular enlargement and/or symptoms, although their systolic pulmonary artery pressure was invariably normal (20-28 mm Hg, median = 25). The angiographic diameter of the defect ranged from 2.5 to 14 mm (median 6 mm).
  
  
  METHODS
  A multi-institutional study.
  
  
  METHODS
  Out of 25 cases attempted, 18 children and adults aged 4-35 years had devices implanted. Fifteen of these patients had membranous ventricular septal defects and three had muscular defects. All patients with a membranous ventricular septal defect had an associated aneurysm of the membranous septum.
  
  
  METHODS
  The buttoned device was introduced either directly or, in the last 12 cases, over a wire bridging the femoral artery and the femoral or jugular vein; the devices were delivered through 7-9 French (F) long sheaths. A membranous defect was regarded as suitable for device closure if the distance from the centre of the defect to the insertion of the right coronary aortic valve leaflet was more than 50% of the size of the required device. The device was guided by echocardiography and fluoroscopy. All muscular defects were corrected through the right jugular vein and all membranous ones through the femoral vein.
  
  
  RESULTS
  All 18 patients underwent initial successful implantation of the device. In thirteen patients the shunts were completely occluded and in the remaining five there were trivial residual shunts. In two patients with membranous ventricular septal defects a change from the original position was noticed at two weeks; mild aortic regurgitation developed in one and the murmur recurred in the other; the devices had to be removed surgically. One patient developed transient third degree atrioventricular block during implantation; no tricuspid regurgitation was observed.
  
  
  CONCLUSIONS
  Clinical occlusion of congenital ventricular septal defects was achieved in 16 out of the 18 attempted cases (13 full occlusions). Membranous ventricular septal defect occlusion can be effective and safe if patients and device sizes are carefully selected.
authors.affiliation: Athenian Institute of Paediatric Cardiology, Athens, Greece.
authors.forename: E B
authors.initials: EB
authors.lastname: Sideris
authors.forename: K P
authors.initials: KP
authors.lastname: Walsh
authors.forename: J L
authors.initials: JL
authors.lastname: Haddad
authors.forename: C R
authors.initials: CR
authors.lastname: Chen
authors.forename: S G
authors.initials: SG
authors.lastname: Ren
authors.forename: H
authors.initials: H
authors.lastname: Kulkarni
mesh_terms: D000293:Adolescent; D000328:Adult; D006328:Cardiac Catheterization; D002648:Child; D002675:Child, Preschool; D004452:Echocardiography; D005500:Follow-Up Studies; D006345:Heart Septal Defects, Ventricular / Q000628:therapy*; D006801:Humans; D008875:Middle Aged; D013523:Surgical Equipment