title: Bipolar leads for use with permanently implantable cardiac pacing systems: a review of limitations of traditional and coaxial configurations and the development and testing of new conductor, insulation, and electrode designs.
abstract: The unacceptable rate of mechanical failures, threshold problems, and recalls experienced with many coaxial bipolar cardiac pacing lead designs are reviewed in detail. To address these problems, redundant insulation coradial atrial and ventricular tined leads (AL and VL, respectively) with iridium oxide electrodes were developed and subjected to extensive accelerated testing. There were no mechanical failures. The new lead body design proved to be much more durable than widely used trifilar MP35N configurations. The data reviewed and early and current test results are strongly supportive of tightly coupled insulation being a major factor in improving lead durability as long as the insulating material is not stressed. In addition to improving flex life, insulation adherence to the conductor may reduce the potential for ionic degradation. Pacing and sensing thresholds in animal studies of the new leads were within the reported range for leads with steroid eluting electrodes. A multicenter Canadian clinical trial was initiated with the first implant in early January 1994. By November 1995, 110 VL and 82 AL had been placed in 124 patients and followed for a mean of 11 +/- 6 months; maximum 21, total 1355. There were 60 males and 64 females with a mean age of 64 +/- 16 years, range 15-88. Primary indications for pacing were AV block in 61 patients, sick sinus syndrome in 53, vasovagal syncope in 4, and congestive heart failure in 7. Many patients had associated or primary tachyarrhythmias, including 111 with supraventricular and 12 with ventricular. Forty-two percent of patients (52/124) had prior cardiac procedures, including 18 open heart surgeries and 20 AV nodal ablations. At implant, 8 lead characteristics were rated good or excellent in 90% (746/829) of evaluations. X-ray visibility was of concern in 10% of patients (12/124). Three perioperative complications occurred, including displacement of one AL (1.2%) and one VL (0.9%). There were no subsequent mechanical (connector, conductor, or insulation) or functional (exit block, micro or macro displacement, or over- or undersensing) problems. Implant pacing thresholds (PT) at 0.45 ms were AL, 0.6 +/- 0.2 (74) and VL 0.4 +/- 0.2 V; impedance (Z) at 3.5 V output AL 373 +/- 77 (82) and VL 497 +/- 117 omega. Sensing thresholds (ST) were AL 3.1 +/- 1.6 (74) and VL 10.3 +/- 4.9 mV. Ventricular lead data were obtained for all patients (N = 110). Atrial lead data are incomplete, because some patients were in atrial fibrillation during implantation. After 12 months, AL PT at 1.5 V output was 0.18 +/- 0.10 ms (21) and at 2.5 V was 0.10 +/- 0.053 (22). Associated AL ST was 3.3 +/- 0.9 mV (21) AL Z 500 +/- 65 omega (25). After 18 months VL PT at 1.5 V was 0.15 +/- 0.10 ms (9) and at 2.5 V output was 0.09 +/- 0.04 ms (9). Associated VL ST was > 7.5 +/- 2.4 mV (9) and VL Z 497 +/- 105 omega (9). Follow-up time discrepancy is due to the VL being available 6 months earlier than the AL. There were no 30-day deaths and only one late death at 10 months in a patient with chronic atrial fibrillation. Death was unrelated to pacer or lead function. At 1 year, 68% AL (15/22) and 62% (24/39) captured at 0.5 V and < or = 1 ms pulse width output. Innovative adherent insulation coradial bipolar lead conductors of the design studied combined with coated iridium oxide electrodes provide for a negligible incidence of mechanical or functional failure with clinical follow-up now approaching 3 years. Excellent acute and chronic sensing and pacing thresholds have been documented. Late thresholds have continued to improve gradually. Long-term clinical pacing at < or = 1.5 V output with a large safety margin is feasible in essentially all patients. This coradial design produces very flexible < 5 French bipolar redundantly insulated lead bodies allowing both AL and VL to simultaneously pass through a single 10 French introducer sheath. (ABSTRACT TRUNCATED)
authors.affiliation: Department of Surgery, University of British Columbia, Vancouver, Canada.
authors.forename: G F
authors.initials: GF
authors.lastname: Tyers
authors.forename: P
authors.initials: P
authors.lastname: Mills
authors.forename: J
authors.initials: J
authors.lastname: Clark
authors.forename: M
authors.initials: M
authors.lastname: Cheesman
authors.forename: J A
authors.initials: JA
authors.lastname: Yeung-Lai-Wah
authors.forename: R R
authors.initials: RR
authors.lastname: Brownlee
mesh_terms: D006348:Cardiac Surgical Procedures; D004566:Electrodes; D004867:Equipment Design / Q000639:trends*; D005069:Evaluation Studies as Topic; D006801:Humans; D008422:Materials Testing; D010138:Pacemaker, Artificial / Q000639:trends*