title: [Cava filter placement under MRI control. Experimental in vitro and in vivo studies].
abstract: OBJECTIVE
  An instrument has been developed for the introduction of caval filters which can be used with MRI; it has been investigated in in vitro and in in vivo experiments.
  
  
  METHODS
  The ferromagnetic components of a commercially available instrument for the femoral introduction of the MR-eye Tulip IVC filter were changed for similar, non-ferromagnetic parts and the lock and dilator marked with dysprosium oxide rings. The instrument was used in a flow phantom and in animal experiments (two domestic pigs) in order to insert filters under MRI control on a 1.5 T Philips Gyroscan with integrated mobile digital subtraction angiography.
  
  
  RESULTS
  Both in vitro and in vivo, the introducer, catheter and caval filter could be identified by MRI and positioned under MRI control. The position of the filter as indicated by MRI corresponded with radiological and macroscopic findings in all cases (5 phantoms, 2 pigs).
  
  
  CONCLUSIONS
  The early experimental results indicate that percutaneous introduction of caval filters with placement under MRI control is possible.
authors.affiliation: Klinik für Radiologische Diagnostik, Universitätsklinikum der RWTH Aachen.
authors.forename: J
authors.initials: J
authors.lastname: Neuerburg
authors.forename: A
authors.initials: A
authors.lastname: Bücker
authors.forename: G
authors.initials: G
authors.lastname: Adam
authors.forename: M
authors.initials: M
authors.lastname: Leenen
authors.forename: E
authors.initials: E
authors.lastname: Rasmussen
authors.forename: M S
authors.initials: MS
authors.lastname: Hurtak
authors.forename: R W
authors.initials: RW
authors.lastname: Günther
mesh_terms: D000818:Animals; D008279:Magnetic Resonance Imaging / Q000295:instrumentation; D008297:Male; D008961:Models, Structural; D019047:Phantoms, Imaging; D011655:Pulmonary Embolism / Q000517:prevention & control; D013552:Swine; D016306:Vena Cava Filters