An accurate method for computer-generating tungsten anode x-ray spectra from 30 to 140 kV.
A tungsten anode spectral model using interpolating polynomials (TASMIP) was used to compute x-ray spectra at 1 keV intervals over the range from 30 kV to 140 kV. The TASMIP is not semi-empirical and uses no physical assumptions regarding x-ray production, but rather interpolates measured constant potential x-ray spectra published by Fewell et al. [Handbook of Computed Tomography X-ray Spectra (U.S. Government Printing Office, Washington, D.C., 1981)]. X-ray output measurements (mR/mAs measured at 1 m) were made on a calibrated constant potential generator in our laboratory from 50 kV to 124 kV, and with 0-5 mm added aluminum filtration. The Fewell spectra were slightly modified (numerically hardened) and normalized based on the attenuation and output characteristics of a constant potential generator and metal-insert x-ray tube in our laboratory. Then, using the modified Fewell spectra of different kVs, the photon fluence phi at each 1 keV energy bin (E) over energies from 10 keV to 140 keV was characterized using polynomial functions of the form phi (E) = a0[E] + a1[E] kV + a2[E] kV2 + ... + a(n)[E] kVn. A total of 131 polynomial functions were used to calculate accurate x-ray spectra, each function requiring between two and four terms. The resulting TASMIP algorithm produced x-ray spectra that match both the quality and quantity characteristics of the x-ray system in our laboratory. For photon fluences above 10% of the peak fluence in the spectrum, the average percent difference (and standard deviation) between the modified Fewell spectra and the TASMIP photon fluence was -1.43% (3.8%) for the 50 kV spectrum, -0.89% (1.37%) for the 70 kV spectrum, and for the 80, 90, 100, 110, 120, 130 and 140 kV spectra, the mean differences between spectra were all less than 0.20% and the standard deviations were less than approximately 1.1%. The model was also extended to include the effects of generator-induced kV ripple. Finally, the x-ray photon fluence in the units of photons/mm2 per mR was calculated as a function of HVL, kV, and ripple factor, for various (water-equivalent) patient thicknesses (0, 10, 20, and 30 cm). These values may be useful for computing the detective quantum efficiency, DQE(f), of x-ray detector systems. The TASMIP algorithm and ancillary data are made available on line at http:/(/)www.aip.org/epaps/epaps.html.
Boone
J M
JM
Department of Radiology, University of California, Davis, Sacramento 95817, USA.
Seibert
J A
JA
D000465:Algorithms; D055592:Biophysical Phenomena; D001703:Biophysics; D003198:Computer Simulation; D006801:Humans; D011789:Quantum Theory; D013057:Spectrum Analysis / Q000706:statistics & numerical data*; D013679:Technology, Radiologic; D014414:Tungsten; D014965:X-Rays
D014414:Tungsten