[Usefulness of magnetic resonance imaging for managing patients with prosthetic carbon valve in the mitral position].
The safety, findings and clinical usefulness of magnetic resonance (MR) imaging were assessed in patients with a prosthetic carbon valve in the mitral position. In vitro deflection, heating and image distortion due to the magnetic field of a 1.5 tesla MR machine were examined in three carbon valves (CarboMedics, St. Jude Medical and Björk-Shiley valves). In vivo MR imaging of the left ventricular horizontal long-axis, vertical long-axis and short-axis views was performed by electrocardiographically synchronized spin echo and field (gradient) echo techniques in eight patients with prosthetic mitral carbon valves, consisting of six CarboMedics valves, one St. Jude Medical valve and one Björk-Shiley valve. No deflection and significant heating was seen in all three valves in vitro. Although little image distortion was shown in the CarboMedics and St. Jude Medical valves, a small distortion toward the frequency encoded direction was seen in the Björk-Shiley valve but caused no difficulty in assessing the surrounding images. Four of the eight patients had normal sinus rhythm and the other four had atrial fibrillation. The prosthetic valves were depicted as signal voids in the images taken by both spin echo and field echo techniques in vivo. Clear structural information with little image distortion of the adjacent tissues of the prosthetic valves were obtained in all patients, although the image of the Björk-Shiley valve which contained stainless steel in the frame had a slightly stronger distortion than those of the CarboMedics and St. Jude Medical valves which contained titanium. The stainless wire suture material used to close the sternal incision was depicted as a signal void, and the areas of the signal loss were larger in the images taken by the field echo technique than those by the spin echo technique. The images taken by the spin echo technique in patients with atrial fibrillation had reduced quality due to the irregularity of repetition time. Cine MR imaging by the field echo technique showed physiological mitral regurgitant jets as signal loss within the flowing blood, which appeared as high signal intensity, bidirectionally in the bileaflet mechanical valve and unidirectionally in the monoleaflet mechanical valve. An abnormal cavity was seen behind the basal left ventricular myocardium in one patient with a CarboMedics valve. The wall of the abnormal cavity was disrupted abruptly and the rest of the wall consisted of pericardium and adjacent tissue in the image taken by the spin echo technique. The image taken by the field echo technique showed an abnormal jet flow from the basal part of the left ventricular cavity into the abnormal cavity, which was compatible with left ventricular pseudoaneurysm. Two-dimensional echocardiography and Doppler color flow mapping disclosed the abnormal cavity and the abnormal flow inside, but failed to show the connection between the left ventricle and the cavity due to reverberation of the ultrasound signal by the prosthetic valve. These findings suggest that MR imaging is a safe and promising method to assess the complications and valvular function in patients with a prosthetic carbon valve in the mitral position.
Koito
H
H
Second Department of Internal Medicine/Cardiovascular Center, Kansai Medical University, Osaka.
Imai
Y
Y
Suzuki
J
J
Ohkubo
N
N
Nakamura
C
C
Takahashi
H
H
Iwasaka
T
T
Inada
M
M
D000368:Aged; D002244:Carbon; D005260:Female; D006349:Heart Valve Diseases / Q000175:diagnosis; D006350:Heart Valve Prosthesis; D019918:Heart Valve Prosthesis Implantation; D006801:Humans; D008279:Magnetic Resonance Imaging / Q000379:methods*; D008297:Male; D008875:Middle Aged; D008943:Mitral Valve / Q000473:pathology*
D002244:Carbon