The effects of electrode-tissue contact on radiofrequency lesion generation.
During the generation of radiofrequency (RF) lesions in the ventricular myocardium, the maintenance of adequate electrode-tissue contact is critically important. In this study, lesion dimensions and temperature and impedance changes were evaluated while controlling electrode-tissue contact levels (-5, 0, +1, and +3 mm) and power levels (10, 20, and 30 W). This data was used to assess the ability of impedance and temperature monitoring to provide useful information about the quality of electrode-tissue contact. The results show that as the electrode-tissue contact increases, so does the amount of temperature rise. With the electrode floating in blood (-5 contact), the average maximum temperature increase with 20 and 30 W was only 7 +/- 1 and 11 +/- 2 degrees C, respectively. At 20 and 30 W the temperature plateaued shortly after the initiation of power application. With good electrode-tissue contact (+1 mm or +3 mm), the temperature increase within the first 10 seconds was significantly greater than the temperature increase from baseline with poor contact (0 mm or -5 mm) and reached a maximum of 60 +/- 1 degrees C after 60 seconds of power application. As the electrode-tissue contact increased, so did the rate and level of impedance decrease. However, the rate of impedance decrease was slower compared to the rate of temperature rise. With the electrode floating in blood, the maximum impedance decreases with 20 and 30 W were 6 +/- 6 omega and 9 +/- 5 omega, respectively. The impedances plateaued after a few seconds of power application. With the electrode in good contact, the maximum impedance decreases with 20 and 30 W were 25 +/- 2 omega and 20 +/- 6 omega, respectively. In these cases the rate of the impedance decrease plateaued after 40 seconds of power application. The increase in lesion diameter and depth correlate well with decreasing impedance and increasing temperature. However, lesion depth appears to correlate better with impedance than temperature. We conclude that, since the electrode-tissue contact is not known prior to the application of power to the endocardium, in the absence of a temperature control system, the power should initially be set at a low level. The power should be increased slowly over 20-30 seconds, and then maintained at its final level for at least 90 seconds to allow for maximal lesion depth maturation. The power level should be lowered if the impedance drop exceeds 15 omega.
Avitall
B
B
Electrophysiology Laboratory, University of Wisconsin-Milwaukee Clinical Campus, Sinai Samaritan Medical Center, USA.
Mughal
K
K
Hare
J
J
Helms
R
R
Krum
D
D
D000818:Animals; D001831:Body Temperature; D017115:Catheter Ablation / Q000295:instrumentation / Q000592:standards; D004285:Dogs; D017097:Electric Impedance; D004566:Electrodes; D006358:Hot Temperature; D008991:Monitoring, Physiologic; D010496:Pericardium / Q000502:physiology / Q000601:surgery*; D011237:Predictive Value of Tests