Incidence of Lightspeed separation and the potential for bypassing.
It is important for endodontic instruments to have a low fracture rate. If a fracture does occur, it would be desirable to have the ability to bypass the broken segment and complete the root canal treatment. One hundred sixty-two root canals in 52 maxillary and mandibular first molars were cleaned and shaped with Lightspeed instruments by three endodontists in their private practices. The canals were instrumented using the technique recommended by the manufacturer. All canals were instrumented to at least a size 45 at the working length. Six instruments separated during treatment. All six had been used more times than recommended by the manufacturer. Five of the six were easily bypassed and treatment completed.
Ramirez-Salomon
M
M
Soler-Bientz
R
R
de la Garza-González
R
R
Palacios-Garza
C M
CM
D000293:Adolescent; D000328:Adult; D004868:Equipment Failure; D019544:Equipment Failure Analysis; D006801:Humans; D008334:Mandible; D008437:Maxilla; D008963:Molar; D018915:Root Canal Preparation / Q000295:instrumentation* / Q000379:methods