Personal thoracic CIP10-T sampler and its static version CATHIA-T.
A specific version of the personal aerosol sampler CIP 10 was designed, named CIP10-T, for sampling the conventional CEN thoracic fraction. A static sampler, named CATHIA, was also designed. It uses the same sampling head, but the size selected particles are collected onto a filter. The combined particle efficiency of the aspiration slot and the selector was measured in a horizontal wind tunnel at low air velocity, close to 16 cm s-1. The flow rate of both samplers was fixed at its nominal value, i.e., 71 min-1. Two different methods were used: the former was based on the Aerodynamic Particle Sizer (TSI); the latter used the measurement of particle size distribution of the collected samples by the Coulter technique. For the CIP10-T sampler, the particle collection efficiency onto the rotating cup was also measured. For both samplers bias and accuracy maps have been calculated, following the recommendations of a new CEN standard about sampler performance. The bias does not exceed 10% in absolute value for both samplers, within a large range of particle size distribution of the total aerosol. For the CIP10-T sampler, the accuracy map exhibits a large area where the accuracy is better than 10%, corresponding for example to 4 microns < or = MMAD < or = 14 microns for GSD = 2. For the same geometric standard deviation, the accuracy is still better than 20% for 15 microns < or = MMAD < or = 21 microns. For the CATHIA-T sampler, the accuracy map can be roughly divided into two parts. The accuracy remains better than 10% for MMAD < or = 12 microns, and it remains between 10 and 20% for coarser aerosols, with 13 microns < or = MMAD < or = 20 microns, provided GSD > or = 2.
Fabriès
J F
JF
Institut National de Recherche et de Sécurité, Vandoeuvre, France.
Görner
P
P
Kauffer
E
E
Wrobel
R
R
Vigneron
J C
JC
D000336:Aerosols; D004867:Equipment Design; D005374:Filtration; D006801:Humans; D016273:Occupational Exposure; D016272:Occupational Health; D010316:Particle Size
D000336:Aerosols