title: [Measurement of glomerular filtration rate (GFR) after administration of iodine contrast medium with the Renalyzer PRX90 in healthy cats and cats with kidney diseases].
abstract: In the present study, the measurement of the glomerular filtration rate (GFR) in the cat with the aid of an iodine contrast medium clearance with the renalyzer PRX90 is introduced. Investigations on the accuracy of measurement showed that even repeated measurement of plasma samples after two days of storage at room temperature yielded reproducible clearance results. Also, partial dilution of the plasma sample (2 ml with 1 ml Aqua bidest.) to reduce the volume of blood withdrawn still produced reliable results. Further dilution of the plasma volume (1 ml with 2 ml Aqua bidest.) however did not allow for accurate measurements. A total of 59 cats of different age and sex were included in the study. 31 cats had healthy kidneys with urea and creatinine values within the reference range, unchanged urine findings and physiologic urine protein patterns (SDS-PAGE). These cats served as reference group. The GFR reference value ascertained for these animals was 2.1 ml/min/kg BW (mean = 3.45 ml/min/kg with s = +/- 1.0 ml/min/kg). 28 cats had elevated values of urea and creatinine in the blood, as well as partially changed urine findings. For further diagnosis of renal disease, separation of urine proteins was done with the SDS-PAGE in the PhastSystem, which in all cases yielded a pathologic urine protein pattern. In 11 cases the renal disease could be further confirmed by histological investigation. GFR in these patients was clearly lowered compared with healthy cats, with measured values between 0 and 1.8 ml/min/kg. It can be concluded that the renalyzer allows reliable determination of the GFR also in the cat. To what extent measurement of the GFR is also helpful to diagnose nephropathies in the stage of compensation needs to be further investigated. In cats with high grade uremia and a GFR below 1 ml/min however, an exact calculation is not possible, since the accuracy of measurement within this range is inadequate. Thus, in severe disease no correct assessment is possible, and no statement concerning prognosis can be made.
authors.affiliation: Klinik für kleine Haustiere, Tierärztlichen Hochschule Hannover.
authors.forename: A
authors.initials: A
authors.lastname: Meyer-Lindenberg
authors.forename: A
authors.initials: A
authors.lastname: Westhoff
authors.forename: P
authors.initials: P
authors.lastname: Wohlsein
authors.forename: J
authors.initials: J
authors.lastname: Pohlenz
authors.forename: I
authors.initials: I
authors.lastname: Nolte
mesh_terms: D000818:Animals; D002371:Cat Diseases / Q000503:physiopathology*; D002415:Cats / Q000502:physiology*; D005260:Female; D005919:Glomerular Filtration Rate; D007674:Kidney Diseases / Q000503:physiopathology / Q000662:veterinary*; D007677:Kidney Function Tests / Q000295:instrumentation*; D008297:Male; D012016:Reference Values