Why and how is soft copy reading possible in clinical practice?
The properties of the human visual system (HVS) relevant to the diagnostic process are described after a brief introduction on the general problems and advantages of using soft copy for primary radiology interpretations. At various spatial and temporal frequencies the contrast sensitivity defines the spatial resolution of the eye-brain system and the sensitivity to flicker. The adaptation to the displayed radiological scene and the ambient illumination determine the dynamic range for the operation of the HVS. Although image display devices are determined mainly by state-of-the-art technology, analysis of the HVS may suggest technical characteristics for electronic displays that will help to optimize the display to the operation of the HVS. These include display size, spatial resolution, contrast resolution, luminance range, and noise, from which further consequences for the technical components of a monitor follow. It is emphasized that routine monitor quality control must be available in clinical practice. These image quality measures must be simple enough to be applied as part of the daily routine. These test instructions might also serve as elements of technical acceptance and constancy tests.
Siemens Medical Engineering Group, Erlangen, Germany.
T
T
Mertelmeier
D000221:Adaptation, Ocular / Q000502:physiology; D016477:Artifacts; D015350:Contrast Sensitivity / Q000502:physiology; D003626:Data Display; D003952:Diagnostic Imaging; D005425:Flicker Fusion / Q000502:physiology; D006801:Humans; D007089:Image Enhancement; D007091:Image Processing, Computer-Assisted / Q000379:methods*; D008027:Light; D011785:Quality Assurance, Health Care; D011873:Radiology Information Systems; D013028:Space Perception / Q000502:physiology; D013679:Technology, Radiologic; D014785:Vision, Ocular / Q000502:physiology; D014796:Visual Perception / Q000502:physiology