title: Body support testing and rating.
abstract: The REF, C, and score are a means to give quantitative values for product performance. Other product variables defined as humidity control factor, shear control factor, differential temperature control factor, product life factor, and cost factor were not addressed as part of this study. Additional variables include delivery, maintenance, and warranties. These components can be added to the score in some weighted manner as they become clinically founded. Interface pressures and shears were not measured for beds that are gatched. Each product tested will have characteristics pertinent to its own design that must be addressed to minimize sacral and coccyx skin breakdown when subjects are in this posture because some pneumatic pads bottom out to give excessive sacral and coccyx pressure while others wrinkle when used in this position. This is of major concern when one considers that the coccyx is more susceptible to pressure than any other bony prominence. There may be some aversion to labeling support product performance because it will allow the purchaser to know what is being bought. This initial study indicates the feasibility of quantitating what all patients require: a valid interface support surface for each specific need. The user can be adequately informed before the purchase as to a product's merits without being influenced by the results of inadequate testing, referrals, or sales media influences. Although cushions, shoes, or other prosthetic/orthotic devices were not part of this study, it is appropriate to rate all these devices in a similar manner by simply comparing pressure relief attained to that desired, expressing the result as a percentage of the worst case, and then labeling the product accordingly. Those subjects with sensation can rate comfort as part of the overall score. For cushions, ischial tuberosity pressure relief can be expressed as a percentage of the maximum attainable relief. Metatarsal head relief during gait as well as during static testing can be measured, and claims can be made for shoes and sneakers. Where impact loads and shear forces are paramount, the score must comprise appropriate parameters. Similarly, amputee distal stump relief as a percentage of worst case could be specified by the fitter. Iliac crest relief as a percentage of worst case can also be specified by the fitter. In all instances the score is not a subjective rating but one that can be measured correctly. All body support products can be given an REF, C value, and score as well as other qualifying values where appropriate instrumentation and sufficient subject sampling is used.
authors: J E
authors: JE
authors: Rogers
mesh_terms: D001513:Beds / Q000592:standards* / Q000706:statistics & numerical data; D001835:Body Weight; D004867:Equipment Design; D006801:Humans; D008422:Materials Testing / Q000379:methods* / Q000706:statistics & numerical data; D011312:Pressure; D003668:Pressure Ulcer / Q000517:prevention & control*; D012107:Research Design; D014481:United States