title: Whole-body plethysmography, does it measure tidal volume of small animals?
abstract: A whole-body plethysmograph was used for mice. The increase in pressure caused by each inhalation was equivalent to the increase that could be calculated to result from heating and humidification of the inhaled air. However, comprehending that a drop in temperature and humidity would cause an abrupt pressure decline during exhalation was difficult. Pressure changes in the plethysmograph were also studied with an artificial chest, modeling the respiratory mechanics, but without the "inhaled" air being heated or humidified. The "chest" consisted of a metal bellows oscillated by a stepper motor 25 to 175 times per minute. Hereby air (0.05 to 0.20 mL) moved in and out of the bellows. The air passed through a polyethylene tube, the length of which was proportional to "airway resistance" and varied from 5 to 35 cm. It was found that the pressure oscillation was affected not only by "tidal volume" of the mechanical chest but also by "respiratory rate" and by "airway resistance." We concur with previous investigators that the plethysmograph pressure reflects alveolar pressure and that fluctuations cannot be explained by changes in temperature and humidity. Accordingly, tidal volume can only be qualitatively and not quantitatively assessed.
authors: Department of Gynecology-Obstetrics, State University of New York at Buffalo, 14222, USA. gee1@acsu.buffalo.edu
authors: G
authors: G
authors: Enhorning
authors: S
authors: S
authors: van Schaik
authors: C
authors: C
authors: Lundgren
authors: I
authors: I
authors: Vargas
mesh_terms: D000818:Animals; D008297:Male; D051379:Mice; D008807:Mice, Inbred BALB C; D010993:Plethysmography, Whole Body; D012123:Pulmonary Ventilation / Q000502:physiology; D012121:Respiration, Artificial / Q000295:instrumentation / Q000379:methods; D012143:Respiratory Physiological Phenomena; D013990:Tidal Volume / Q000502:physiology*; D012122:Ventilators, Mechanical