[Capsular bent ring. Improvement of the capsular tension ring].
Since the first introduction of the capsular ring in 1991, the open capsular tension ring proved to be efficient in various indications and applications: e.g. in zonular dialysis or or weakness, with integrated tinted sector shield in sector or aniridia, to facilitate primary posterior capsulorhexis, for prophylaxis of intraocular lens decentration or as an intraocular measuring gauge. Nishi modified this capsular tension ring mainly by increasing the width to 0.7 mm and created a rectangular edge, thus facilitating the creation of a sharp, discontinuous bend in the equatorial capsule. This band-shaped, sharp-edged capsule-bending ring prevented the anterior capsule from coming into contact with the posterior capsule. This is most likely due to the 0.7 mm width of the capsule-bending ring, which significantly prevented anterior capsule opacification, which commonly occurs within one month postoperatively. Posterior capsule opacification seems to be reduced as well, but final evaluation is needed for a longer period of time. The capsular tension ring as well as the capsular bending ring enlarge our surgical options in modern cataract surgery. Both can be considered as a useful instrument and promising innovation.
Dick
B
B
Universitäts-Augenklinik Mainz.
Schwenn
O
O
D000368:Aged; D000369:Aged, 80 and over; D002386:Cataract / Q000209:etiology / Q000473:pathology / Q000517:prevention & control*; D002387:Cataract Extraction / Q000295:instrumentation*; D002675:Child, Preschool; D004867:Equipment Design; D005260:Female; D005500:Follow-Up Studies; D006801:Humans; D007903:Lens Capsule, Crystalline / Q000473:pathology; D007910:Lenses, Intraocular; D008297:Male; D008875:Middle Aged; D011183:Postoperative Complications / Q000209:etiology / Q000473:pathology / Q000517:prevention & control*